Dose reduction in whole-body computed tomography of multiple injuries (DoReMI): protocol for a prospective cohort study
- Dirk Stengel†1, 2, 4Email author,
- Caspar Ottersbach†1,
- Thomas Kahl3,
- Constanze Nikulka3,
- Claas Güthoff1,
- Thomas Hartel2,
- Sophia Hünnebeck1, 2,
- Axel Ekkernkamp2 and
- Sven Mutze3
© Stengel et al.; licensee BioMed Central Ltd. 2014
Received: 26 December 2013
Accepted: 22 February 2014
Published: 3 March 2014
Single-pass, contrast-enhanced whole body multidetector computed tomography (MDCT) emerged as the diagnostic standard for evaluating patients with major trauma. Modern iterative image algorithms showed high image quality at a much lower radiation dose in the non-trauma setting. This study aims at investigating whether the radiation dose can safely be reduced in trauma patients without compromising the diagnostic accuracy and image quality.
Prospective observational study with two consecutive cohorts of patients.
Setting: A high-volume, academic, supra-regional trauma centre in Germany.
Study population: Consecutive male and female patients who 1. had been exposed to a high-velocity trauma mechanism, 2. present with clinical evidence or high suspicion of multiple trauma (predicted Injury Severity Score [ISS] ≥16) and 3. are scheduled for primary MDCT based on the decision of the trauma leader on call.
Imaging protocols: In a before/after design, a consecutive series of 500 patients will undergo single-pass, whole-body 128-row multi-detector computed tomography (MDCT) with a standard, as low as possible radiation dose. This will be followed by a consecutive series of 500 patients undergoing an approved ultra-low dose MDCT protocol using an image processing algorithm.
Data: Routine administrative data and electronic patient records, as well as digital images stored in a picture archiving and communications system will serve as the primary data source. The protocol was approved by the institutional review board.
Main outcomes: (1) incidence of delayed diagnoses, (2) diagnostic accuracy, as correlated to the reference standard of a synopsis of all subsequent clinical, imaging, surgical and autopsy findings, (3) patients’ safety, (4) radiation exposure (e.g. effective dose), (5) subjective image quality (assessed independently radiologists and trauma surgeons on a 100-mm visual analogue scale), (6) objective image quality (e.g., contrast-to-noise ratio).
Analysis: Multivariate regression will be employed to adjust and correct the findings for time and cohort effects. An exploratory interim analysis halfway after introduction of low-dose MDCT will be conducted to assess whether this protocol is clearly inferior or superior to the current standard.
Although non-experimental, this study will generate first large-scale data on the utility of imaging-enhancing algorithms in whole-body MDCT for major blunt trauma.
Current Controlled Trials ISRCTN74557102.
According to the most recent injury statistics, every two minutes someone dies of an injury in the European Union . Trauma is, and will remain a significant contributor to years of productive life lost, disability, and health care service costs worldwide [2–4].
The survival and functional prognosis of patients who suffered severe and multiple injuries improved significantly in the industrialized countries [5–8]. Time to appropriate care is an accepted predictor of trauma outcomes, and transport, triage, and work-up intervals must respect this decisive factor. Also, immediately life-threatening injuries and sources of bleeding must be identified (or excluded) with high accuracy [9, 10]. Apart from damage-control resuscitation and surgery, the introduction of contrast-enhanced, whole-body, multi-detector computed tomography may be regarded as one of the most important interventions which changed the face of trauma care during the past decade [11–14].
In a recent survey of UK emergency departments, 41 out of 184 (22.3%) respondents indicated that they had a pan-scan policy for major trauma patients, with marked regional variations in availability and reporting times . Data from the German TraumaRegisterD suggest that incorporating a routine pan-scan into trauma resuscitation may increase the ratio of observed to expected survivors . A recent analysis confirmed the results in a subsequent cohort and unstable patients . Yet, many injuries to solid organs (e.g., brain, liver, spleen) demarcate only after sufficient resuscitation with crystalloid fluids and blood products. We recently demonstrated in a large cohort of 982 patients that an initial pan-scan is capable to prove but not to exclude injuries to various body regions .
Exposure to radiation for diagnostic purposes is an emerging ethical, medico-legal and public-health issue . It has been estimated that, with the widespread and liberal use of CT, 1.5 - 2.0% of all cancers in the US may now be attributable to the radiation from CT examinations . The median lifetime attributable risk (LAR) of cancer is 4.0 (IQR 0.8 - 11.1) / 1000 for a multiphase abdomen and pelvis CT scan .
Modern 64-, 128- and 256-row MDCT scanners are regarded to produce lower dose indexes (CTDIvol) than earlier generation single- or 4-row hardware. However, Harrieder et al. recently demonstrated in a retrospective study that, because of enhanced scan lengths, the dose-length-product (DLP) may remain similar or even increase with modern hardware . The authors, however, provided little information on patient demographics (e.g., BMI, ISS) and did not only compare different multi-detector-row technologies but also scanners made by different manufacturers. A comparison of 64-row scanners of four leading companies suggested highest patient mean effective doses (in mSv) produced by the GE LightSpeed VCT (12.7 ± 2.6), followed by the Toshiba Aquilion (11.1 ± 3.3), Philips Brilliance (9.5 ± 0.4) and Siemens Somatom (9.1 ± 1.1) .
So-called adaptive statistical iterative algorithms (ASIR), derived from a Bayesian framework, incorporate i. random fluctuations in sinogram measurements , ii. non-ideality, iii. different degrees of data credibility, iv. a priori information about the distribution of the image space, and other previously unconsidered variables to enhance resolution and reduce artefacts and noise .
Distinct technologies like Veo (Model Based Image Reconstruction, GE Healthcare) or IRIS (Iterative Reconstruction in Image Space, Siemens) received FDA approval .
Apart from phantom studies and smaller case series, there is currently sparse evidence from head-to-head studies about the effect of ASIR technology on objective and subjective image quality, and exposure to radiation. Exemplarily, Prakash et al. from Boston showed in a cohort of 222 patients who had previously undergone abdominal CT with filtered back projection (FBP) technology and were scheduled for follow-up ASIR scan that CTDLvol and DLP were markedly reduced without compromising image quality . The Prakash group showed similar results in chest CT [33, 34]. Other investigators reported similar dose reductions with similar image quality in CT portovenography , cerebral CT , and abdomino-pelvic CT scans .
There is evidence from phantom tests and rigorous clinical evaluations with global clinical collaborators which demonstrate the potential of the FDA-approved iDose system (Philips Healthcare, Eindhoven, The Netherlands) to improve image quality and / or lower radiation dose levels beyond those previously achievable with conventional, routine-dose acquisitions, or FBP reconstructions [38–43].
The combination of BMI-adapted protocols with iterative reconstruction algorithms can reduce radiation exposure to patients and simultaneously improve image quality .
Previous non-trauma / elective CT investigations suggest that iterative reconstruction algorithms may maintain image quality at a much lower radiation dose. Yet, large-scale, prospective evidence on the safety and effectiveness of low-dose CT in trauma patients is still sparse. A single retrospective study investigated possible dose reduction at maintained image quality in subjects with multiple injuries . Since results were not adjusted for baseline imbalances (e.g., a 7% difference in the prevalence of comorbidity), conclusions and inferences from this study are difficult to interpret.
Patients with multiple injuries are typically young and thus at a higher risk of CT-induced cancer. Reducing exposure to radiation with constant image quality and diagnostic accuracy in this vulnerable cohort is a key goal of trauma services worldwide, and evidence whether imaging at lower doses of radiation is feasible, safe, and effective is needed. On the other hand, it may be argued that, given the proven benefit of MDCT in lowering trauma-associated mortality, the primary scan must exhaust the full potential of the method and should aim for maximum quality of images to avoid false-negative results, regardless of exposure to radiation.
The Dose Reduction in whole-body computed tomography of Multiple Injuries (DoReMI) investigation is a prospective cohort study employing two large, consecutive cohorts of patients undergoing different approved and established MDCT protocols. All data will be recorded, stored and processed under the conditions of daily practice of trauma care. Gathered data will form the basis for a later interrupted time-series comparison between the effectiveness and safety of the scanning protocols.
Design history and IRB concerns
The study was originally planned as non-inferiority randomized controlled trial (NI-RCT) with the following hypotheses:
Null hypothesis H0 notated as:
where C = image quality in the standard-of-care arm, T = image quality in the experimental (iDose) arm, and M = non-inferiority margin
Alternative hypothesis HA expressed as
In clinical terms, the primary hypothesis read as:
In patients with multiple trauma scheduled for single-pass, contrast-enhanced whole-body 128-row MDCT, the iDose algorithm will generate a subjective image quality rated on a visual analogue scale (0 – 100 mm) which is not inferior to standard image production and processing.
The non-inferiority margin (NIM) was set at 1 mm, with a standard deviation of 4 mm (derived from a pilot study of 2 × 20 patients). The sample size for a non-inferiority trial of two means was computed using PASS 11.0 software (NCSS, LLC. Kaysville, Utah, USA). For a fixed NIM of 1 mm, the needed sample size per group was 253 patients to confirm non-inferiority of the experimental compared to the control intervention with a one-sided alpha 0.025 and a power of 80%.
The protocol was submitted to the institutional review board (IRB) of the Charité University Medical Centre, Berlin, Germany, in May 2013. It accounted for 1. the inclusion of unconscious and minor patients in an emergency setting (with distinct processes of informing relatives and deferred informed consent procedures) and 2. varying exposure to radiation for diagnostic purposes as the intervention of interest. Given that MDCT trauma imaging at a lower than usual radiation dose is not the current standard of care in Germany, the institutional review board issued concerns that patients in the experimental group are at risk of missed injuries (not that patients in the standard-of-care group may be exposed to a higher than needed effective dose). Consequently, the initial proposal was rejected. A revised proposal submitted in June 2013 could not solve the IRB’s concerns.
Consequently, the investigators changed their approach to comply with the IRB’s recommendations, and consented on a before/after cohort study. All patients in this study will receive an accepted diagnostic standard of care using approved technologies but will not undergo random allocation to either modality. One may still argue that injuries are acute, unforeseeable events, and there remains a random component in the study set-up.
The ultimate consequence of shifting from an experimental to an observational design was the increase in the sample size to allow for robust multivariate modelling and adjustment of results for confounding.
To determine the incidence of missed or delayed diagnoses with either imaging scheme,
To define the diagnostic accuracy of either MDCT protocol, based on the synopsis of clinical, imaging, and interventional information gathered until discharge,
To evaluate patients’ safety with either protocol,
To determine the exposure to radiation and the LAR of cancer with either approach,
To evaluate subjective and objective measures of image quality.
Key scanning parameters in the experimental and the control group
128-row MDCT pan-scan
128-row MDCT pan-scan + iDose
Detector z coverage, mm
Delayed diagnoses comprise pathological anatomic and / or physiologic findings which demand clinical awareness, or, in case of therapeutically relevant pathological results, should prompt immediate therapeutic action.
We will estimate the incidence of delayed diagnoses based on a synopsis of all subsequent clinical, imaging, intra-operative, and autopsy findings until discharge. We will further distinguish between clinically relevant and all diagnoses. Relevant diagnoses are defined as severe injuries suiting an Abbreviated Injury Scale [AIS] score ≥3 and/or injuries demanding surgical and/or trans-vascular intervention. Non-traumatic incidental findings like tumours or tumour-like lesions demanding specific work-up and/or treatment will be recorded in an exploratory attempt.
The general technical capability of CT technology to depict a certain pathology (resolution)
Trauma-associated, acutely impaired physiological parameters (e.g., hypotension, haemorrhagic shock, centralized circulation) which hamper organ perfusion and/or vessel contrast
The interval needed for tissue contusions and hematomas to demark or expand
Interpretation of images by the radiologist on call
The severity and/or therapeutic consequences of findings
In addition to the incidence of delayed diagnoses, we will compute standard indicators of diagnostic test accuracy (i.e., sensitivity, specificity, positive and negative predictive values, areas under the ROC). Different models will be created for various endpoints (e.g., all missed injuries, therapeutically relevant findings, and others).
Any situation in which the individual patient’s safety is assumed to be compromised will be recorded and evaluated in a cumulative fashion. Safety-relevant events will be entered in the local Critical Incident Reporting System (CIRS), hosted by the Medical Council (Ärztekammer) of Berlin, Germany. Since this is an observational study, we will use common definitions of adverse events (AE), serious adverse events (SAE), and Suspected Unexpected Serious Adverse Reactions (SUSAR), as outlined in ICH Topic E2B (Clinical Safety Data Management: Data Elements for Transmission of Individual Case Safety Reports), and most recently in the amended and updated DIN-ISO 14155 guideline, but do not implement a regular reporting procedure.
Radiation dose and estimated LAR of cancer
CTDIvol, scan length, DLP as reported by the scanner (dose report page in PACS), and verified by software CT-Expo, if values are ambiguous or information is missing
Effective dose (in mSv) calculated by CT-Expo from documented parameters, based on most recent ICRP recommendations (modified tissue-weighting factors)
Subjective image quality
Numerous approaches have been proposed to quantify the quality of CT images in the transversal plane, and to generate resource-intensive but clinically required coronal, sagittal, and 3-D reconstruction views.
Subjective evaluation of quality, ideally assessed by two or more independent expert raters in a random, blinded (towards the underlying processing algorithm) fashion, using pre-defined body areas and instruments or scales
Objective evaluation, using physical parameters like contrast/signal, noise, and the ratio of both
There is currently no consensus or a reproducible pattern of criteria available to distinguish “good” or “diagnostic” from “unacceptable” or “un-diagnostic” images. In 1999, the European Commission had summarized quality criteria in a guideline (EUR 16262 EN, http://w3.tue.nl/fileadmin/sbd/Documenten/Leergang/BSM/European_Guidelines_Quality_Criteria_Computed_Tomography_Eur_16252.pdf) which had lastly been updated in 2004 (http://www.msct.eu/CT_Quality_Criteria.htm).
Latest updates on the web-platform of the Imaging Performance Assessment of CT Scanners (IMPACT, http://www.impactscan.org/) working group have been made in 2010.
From a statistical point of view, 3- and 5-point ordinal scales have the disadvantage of a non-normal / non-parametric distribution. Although frequently used, the central measure of these scales is not the parametric mean but the geometric mean or median. For sample size calculations and full-scale variance, a visual analogue scale is the better option to determine image quality.
To overcome the limitations with established rating scales, wet set up a pilot study using 2 x 20 sets of images from patients with multiple trauma. The control set consisted of images obtained by a 64-row MDCT scanner (Brilliance 64, Philips, Germany), with scanning parameters published elsewhere . The experimental set comprised images obtained by the 128-row MDCT (Ingenuity Core 128, Philips) scanner to be used in the planned project.
Two independent radiologists blinded to the scanner generation and protocol rated image quality on a 100-mm visual analogue scale, focusing on lung, bone, vessel, fluid/fat and organ tissue contrast.
The arithmetic mean of both raters with the pooled standard deviation s, calculated as
was used for quantitative assessment of differences between groups. Here, n1 and n2 are the number patients / images assessed by rater 1 and 2, and S1 and S2 are the standard deviations of means.
Summary of the results of the pilot study
Pooled mean (SD)
Mean difference (95% CI)
0.60 (-0.66 to 1.86)
1.13 (-0.82 to 3.08)
1.70 (-0.95 to 4.35)
3.62 (-0.36 to 7.60)
2.20 (-0.33 to 4.73)
Objective image quality
Contrast and noise, and the ratio of both, are amongst the most influential indices to describe the objective quality of CT images.
where S1 = mean CT number (in HU) in region of interest 1 (ROI1) 1, S2 = mean HU in ROI2, σ1 = mean SD of the mean HU in ROI1, and σ2 mean SD of the mean HU in ROI2
Standard ROIs of similar size and position are placed in prominent anatomical structures such as the aorta, liver, other solid organs and spinal muscles. If metal implants are present, then ROIs shall also be placed in the vicinity.
sustained blunt major trauma by a car crash or fall from a height
been resuscitated on scene by a multi-professional team of paramedics and emergency physicians, including sedation / general anaesthesia and airway management by oro-tracheal intubation
Thus, the target sample unavoidably includes incompetent (i.e., unconscious and ventilated, probably hemodynamically unstable) patients.
An injury is an unpredictable event that may occur in all age groups, and regardless of other medical conditions. Some patients may also not have reached the legal age of consent at the time of injury.
Male and female patients of all ages
Suspected or already proven blunt multiple trauma, as indicated by the presence of injuries to ≥2 body regions which, alone or in combination, are life-threatening, or an ISS ≥16
Indication for primary, single-pass whole-body CT by judgment of the trauma leader in charge and according to red flag criteria outlined in the German Trauma Association’s clinical practice guideline on the management of multiple trauma :
any injury mechanism exposing patients to a high risk of multiple trauma (i.e., road traffic crash with presumed high-energy trauma like extrication or death of a car occupant, pedestrian struck by a vehicle, fall from height)
need for technical rescue
impaired physical or physiological status (i.e., unconsciousness, intubation and ventilation, obvious signs of injury like bruises, haematoma, open wounds or fractures, hemodynamic instability)
suspicion of severe trauma confirmed by paramedics and/or emergency doctors on scene
considered unsuitable for CT for any reason (e.g., need for immediate life-saving thoracotomy, laparotomy, or cranial trepanation)
with futile resuscitation and / or CPR efforts
declared dead on arrival will be missed by this study.
Recruitment and documentation
Patients will be recruited under routine clinical care conditions, and documentation will mainly comprise data collected for administrative issues or routinely entered in the electronic patient chart. Also, all radiological images will be stored in a PACS for later processing and evaluation.
We will record all patients participating in this study immediately after they underwent the MDCT scan. Informed consent will be gathered by relatives or the next of kin, or individual patients as soon as they are able to consent in using their data for scientific purposes.
This study focuses on delayed and missed injuries, and the accuracy of the primary MDCT scan. Thus, it needs a diagnostic reference or gold standard.
The probably best reference method to which initial MDCT findings can be compared is the clinical and radiological follow-up. If the initial MDCT misses injuries in severely injured patients, it is very likely they will be detected during hospital stay (the same holds true for suspected pathological findings which cannot be confirmed or verified on follow-up imaging).
This study will employ a synopsis of all clinical (including intra-operative) and radiological findings collected until discharge as the diagnostic reference standard. All these items will be traced from routinely available information.
Sample size considerations and calculations
The sample size is calculated to guarantee a certain precision of a range of estimates, not to achieve a certain power to detect an a priori defined effect size.
95% confidence intervals of both risk and mean differences must not exceed the point estimate by more than 5% in either direction
multivariate modelling must be possible for five independent covariates (respecting Harrell’s rule)
Accepted statistical methods will be used to determine whether the distribution of baseline variables and outcome results are compatible with chance. In general, results will be presented as means, medians, or proportions, and differences in means and proportions, including appropriate measures of distribution and 95% confidence intervals.
We will use established methods to model observational, specifically time-series data, stratified for the different imaging protocols and potential confounding, independent variables. This will include multi-variable generalized linear, random- and mixed-effects regression analyses.
Missing values will be addressed by adequate methods (including multiple imputation techniques).
Multivariate modelling will always be conducted using sound selection, entry, and exclusion of variables, and after testing for model fit and plausibility.
EQUATOR-, GCP-, GEP-, and AMA-conformity will be assured when presenting results. Raw numbers and counts will be provided together with percentages. The STATA 11.0 software package will be employed for all analyses.
Data management and quality assurance
Data management, including data entry, plausibility checks and query generation will be performed by the local trial coordinating unit (Centre of Clinical Research, Unfallkrankenhaus Berlin).
Subjective quality of images, as assessed on 100-mm VAS for general impression and distinct quality of structures, tissues and sites specified earlier, will be graded by two radiologists in a paper-based fashion. Tick-marks on the VAS will be measured and be entered in an electronic data capture (EDC) system by appointed research assistants at the trial unit.
All further data will be handled and stored electronically for further processing, and processed in the database. Data handling and protection will conducted according to GCP and federal laws of data protection.
All materials pertaining to the investigation will be documented by research staff of the trial unit, sorted and kept in closed archives. The investigator should maintain all source data together with related study documentation for the maximum period of time permitted by the hospital, institution or private practice, but not less than that minimally prescribed by the local authorities after the clinical part of the study has been completed.
This is the first study which assesses the potential value of a dose-reducing MDCT regimen in patients with multiple and severe trauma. The investigators initially planned a RCT which was rejected by the local IRB.
The current protocol aims at balancing regulatory and methodological issues. While the present study (because of its non-experimental design) cannot formally prove that low-dose MDCT is feasible, safe, and evenly accurate compared to normal-dose MDCT in the trauma setting, the cohort size, the setting, and the anticipated characteristics of patient cohorts will guarantee both high internal and external validity of findings. Advanced statistical approaches will allow for quasi-confirmatory inferences. In the absence of large-scales RCTs, this study may provide the best available evidence to decide in favour of one or the other diagnostic approach.
- 1. European Association for Injury Prevention and Safety Promotion: Injuries in the European Union. Report on Injury Statistics 2008–2010. 2013, Amsterdam: EuroSafeGoogle Scholar
- Mathers CD, Loncar D: Projections of global mortality and burden of disease from 2002 to 2030. PLoS Med. 2006, 3: e442-10.1371/journal.pmed.0030442.PubMed CentralView ArticlePubMedGoogle Scholar
- Murray CJ, Vos T, Lozano R, Naghavi M, Flaxman AD, Michaud C, Ezzati M, Shibuya K, Salomon JA, Abdalla S, Aboyans V, Abraham J, Ackerman I, Aggarwal R, Ahn SY, Ali MK, Alvarado M, Anderson HR, Anderson LM, Andrews KG, Atkinson C, Baddour LM, Bahalim AN, Barker-Collo S, Barrero LH, Bartels DH, Basáñez MG, Baxter A, Bell ML, Benjamin EJ: Disability-adjusted life years (DALYs) for 291 diseases and injuries in 21 regions, 1990–2010: a systematic analysis for the Global Burden of Disease Study 2010. Lancet. 2012, 380: 2197-2223. 10.1016/S0140-6736(12)61689-4.View ArticlePubMedGoogle Scholar
- Peden M: UN General Assembly calls for decade of action for road safety. Inj Prev. 2010, 16: 213-10.1136/ip.2010.027839.View ArticlePubMedGoogle Scholar
- Celso B, Tepas J, Langland-Orban B, Pracht E, Papa L, Lottenberg L, Flint L: A systematic review and meta-analysis comparing outcome of severely injured patients treated in trauma centers following the establishment of trauma systems. J Trauma. 2006, 60: 371-378. 10.1097/01.ta.0000197916.99629.eb.View ArticlePubMedGoogle Scholar
- Davidson GH, Hamlat CA, Rivara FP, Koepsell TD, Jurkovich GJ, Arbabi S: Long-term survival of adult trauma patients. JAMA. 2011, 305: 1001-1007. 10.1001/jama.2011.259.View ArticlePubMedGoogle Scholar
- Mackenzie EJ, Rivara FP, Jurkovich GJ, Nathens AB, Frey KP, Egleston BL, Salkever DS, Scharfstein DO: A national evaluation of the effect of trauma-center care on mortality. N Engl J Med. 2006, 354: 366-378. 10.1056/NEJMsa052049.View ArticlePubMedGoogle Scholar
- National Confidential Enquiry into Patient Outcome and Death: Trauma: Who Cares?. 2007, London: National Confidential Enquiry into Patient Outcome and Death, 2007Google Scholar
- Chen CW, Chu CM, Yu WY, Lou YT, Lin MR: Incidence rate and risk factors of missed injuries in major trauma patients. Accid Anal Prev. 2011, 43: 823-828. 10.1016/j.aap.2010.11.001.View ArticlePubMedGoogle Scholar
- Pfeifer R, Pape HC: Missed injuries in trauma patients: a literature review. Patient Saf Surg. 2008, 2: 20-10.1186/1754-9493-2-20.PubMed CentralView ArticlePubMedGoogle Scholar
- Kanz KG, Korner M, Linsenmaier U, Kay MV, Huber-Wagner SM, Kreimeier U, Pfeifer KJ, Reiser M, Mutschler W: Prioritätenorientiertes Schockraummanagement unter Integration des Mehrschichtspiralcomputertomographen. Unfallchirurg. 2004, 107: 937-944. 10.1007/s00113-004-0845-4.View ArticlePubMedGoogle Scholar
- Salim A, Sangthong B, Martin M, Brown C, Plurad D, Demetriades D: Whole body imaging in blunt multisystem trauma patients without obvious signs of injury: results of a prospective study. Arch Surg. 2006, 141: 468-473. 10.1001/archsurg.141.5.468.View ArticlePubMedGoogle Scholar
- Weninger P, Mauritz W, Fridrich P, Spitaler R, Figl M, Kern B, Hertz H: Emergency room management of patients with blunt major trauma: evaluation of the multislice computed tomography protocol exemplified by an urban trauma center. J Trauma. 2007, 62: 584-591. 10.1097/01.ta.0000221797.46249.ee.View ArticlePubMedGoogle Scholar
- Wurmb TE, Fruhwald P, Hopfner W, Roewer N, Brederlau J: Whole-body multislice computed tomography as the primary and sole diagnostic tool in patients with blunt trauma: searching for its appropriate indication. Am J Emerg Med. 2007, 25: 1057-1062. 10.1016/j.ajem.2007.03.016.View ArticlePubMedGoogle Scholar
- Smith CM, Mason S: The use of whole-body CT for trauma patients: survey of UK emergency departments. Emerg Med J. 2011, 29: 630-634.View ArticlePubMedGoogle Scholar
- Huber-Wagner S, Lefering R, Qvick LM, Korner M, Kay MV, Pfeifer KJ, Reiser M, Mutschler W, Kanz KG, Working Group on Polytrauma of the German Trauma Society: Effect of whole-body CT during trauma resuscitation on survival: a retrospective, multicentre study. Lancet. 2009, 373: 1455-1461. 10.1016/S0140-6736(09)60232-4.View ArticlePubMedGoogle Scholar
- Huber-Wagner S, Biberthaler P, Haberle S, Wierer M, Dobritz M, Rummeny E, van Griensven M, Kanz KG, Lefering R, TraumaRegister DGU: Whole-body CT in haemodynamically unstable severely injured patients - a retrospective, multicentre study. PLoS One. 2013, 8: e68880-10.1371/journal.pone.0068880.PubMed CentralView ArticlePubMedGoogle Scholar
- Stengel D, Ottersbach C, Matthes G, Weigeldt M, Grundei S, Rademacher G, Tittel A, Mutze S, Ekkernkamp A, Frank M, Schmucker U, Seifert J: Accuracy of single-pass whole-body computed tomography for detection of injuries in patients with major blunt trauma. CMAJ. 2012, 2012: 2012-Google Scholar
- Asha S, Curtis KA, Grant N, Taylor C, Lo S, Smart R, Compagnoni K: Comparison of radiation exposure of trauma patients from diagnostic radiology procedures before and after the introduction of a panscan protocol. Emerg Med Australas. 2012, 24: 43-51. 10.1111/j.1742-6723.2011.01504.x.View ArticlePubMedGoogle Scholar
- Baumann BM, Chen EH, Mills AM, Glaspey L, Thompson NM, Jones MK, Farner MC: Patient perceptions of computed tomographic imaging and their understanding of radiation risk and exposure. Ann Emerg Med. 2011, 58: 1-7. 10.1016/j.annemergmed.2010.10.018.View ArticlePubMedGoogle Scholar
- Brenner DJ, Hall EJ: Computed tomography–an increasing source of radiation exposure. N Engl J Med. 2007, 357: 2277-2284. 10.1056/NEJMra072149.View ArticlePubMedGoogle Scholar
- Korley FK, Pham JC, Kirsch TD: Use of advanced radiology during visits to US emergency departments for injury-related conditions, 1998–2007. JAMA. 2010, 304: 1465-1471. 10.1001/jama.2010.1408.View ArticlePubMedGoogle Scholar
- Sharma OP, Oswanski MF, Sidhu R, Krugh K, Culler AS, Stombaugh HA, Lauer SK: Radiation trends in trauma patients. Am Surg. 2011, 77: 193-197.PubMedGoogle Scholar
- Snyder GE: Whole-body imaging in blunt multisystem trauma patients who were never examined. Ann Emerg Med. 2008, 52: 101-103. 10.1016/j.annemergmed.2007.03.023.View ArticlePubMedGoogle Scholar
- Tien HC, Tremblay LN, Rizoli SB, Gelberg J, Spencer F, Caldwell C, Brenneman FD: Radiation exposure from diagnostic imaging in severely injured trauma patients. J Trauma. 2007, 62: 151-156. 10.1097/TA.0b013e31802d9700.View ArticlePubMedGoogle Scholar
- Smith-Bindman R, Lipson J, Marcus R, Kim KP, Mahesh M, Gould R, Berrington de González A, Miglioretti DL: Radiation dose associated with common computed tomography examinations and the associated lifetime attributable risk of cancer. Arch Intern Med. 2009, 169: 2078-2086. 10.1001/archinternmed.2009.427.View ArticlePubMedGoogle Scholar
- Harrieder A, Geyer LL, Korner M, Deak Z, Wirth S, Reiser M, Linsenmaier U: Evaluation of radiation dose in 64-row whole-body CT of multiple injured patients compared to 4-row CT. Rofo. 2012, 184: 443-449.View ArticlePubMedGoogle Scholar
- Colli V, Mangini M, Strocchi S, Lumia D, Canì A, Boffano C, Cesarani F, Rizzi E, Ferrara RM, Andreis IA, Bruno J, Gozzi G, Cacciatori M, Ostinelli A, Vite C, Conte L, Fugazzola C: Performance assessment of four 64-slice computed tomographic devices for a typical clinical protocol. J Comput Assist Tomogr. 2011, 35: 57-64. 10.1097/RCT.0b013e3181f124fe.View ArticlePubMedGoogle Scholar
- Thibault JB, Sauer KD, Bouman CA, Hsieh J: A three-dimensional statistical approach to improved image quality for multislice helical CT. Med Phys. 2007, 34: 4526-4544. 10.1118/1.2789499.View ArticlePubMedGoogle Scholar
- Silva AC, Lawder HJ, Hara A, Kujak J, Pavlicek W: Innovations in CT dose reduction strategy: application of the adaptive statistical iterative reconstruction algorithm. AJR Am J Roentgenol. 2010, 194: 191-199. 10.2214/AJR.09.2953.View ArticlePubMedGoogle Scholar
- Kaplan DA: Iterative reconstruction in CT evolves for lower dose, increased clarity. 2012,http://www.diagnosticimaging.com/ct/iterative-reconstruction-ct-evolves-lower-dose-increased-clarity,Google Scholar
- Prakash P, Kalra MK, Kambadakone AK, Pien H, Hsieh J, Blake MA, Sahani DV: Reducing abdominal CT radiation dose with adaptive statistical iterative reconstruction technique. Invest Radiol. 2010, 45: 202-210. 10.1097/RLI.ob013e3181dzfeec.View ArticlePubMedGoogle Scholar
- Prakash P, Kalra MK, Ackman JB, Digumarthy SR, Hsieh J, Do S, Shepard JA, Gilman MD: Diffuse lung disease: CT of the chest with adaptive statistical iterative reconstruction technique. Radiology. 2010, 256: 261-269. 10.1148/radiol.10091487.View ArticlePubMedGoogle Scholar
- Prakash P, Kalra MK, Digumarthy SR, Hsieh J, Pien H, Singh S, Gilman MD, Shepard JA: Radiation dose reduction with chest computed tomography using adaptive statistical iterative reconstruction technique: initial experience. J Comput Assist Tomogr. 2010, 34: 40-45. 10.1097/RCT.0b013e3181b26c67.View ArticlePubMedGoogle Scholar
- Matsuda I, Hanaoka S, Akahane M, Sato J, Komatsu S, Inoh S, Kiryu S, Yoshioka N, Ino K, Ohtomo K: Adaptive statistical iterative reconstruction for volume-rendered computed tomography portovenography: improvement of image quality. Jpn J Radiol. 2010, 28: 700-706. 10.1007/s11604-010-0490-5.View ArticlePubMedGoogle Scholar
- Ren Q, Dewan SK, Li M, Li J, Mao D, Wang Z, Hua Y: Comparison of adaptive statistical iterative and filtered back projection reconstruction techniques in brain CT. Eur J Radiol. 2012, 81: 2597-2601. 10.1016/j.ejrad.2011.12.041.View ArticlePubMedGoogle Scholar
- Hara AK, Paden RG, Silva AC, Kujak JL, Lawder HJ, Pavlicek W: Iterative reconstruction technique for reducing body radiation dose at CT: feasibility study. AJR Am J Roentgenol. 2009, 193: 764-771. 10.2214/AJR.09.2397.View ArticlePubMedGoogle Scholar
- Noel PB, Fingerle AA, Renger B, Munzel D, Rummeny EJ, Dobritz M: Initial performance characterization of a clinical noise-suppressing reconstruction algorithm for MDCT. AJR Am J Roentgenol. 2011, 197: 1404-1409. 10.2214/AJR.11.6907.View ArticlePubMedGoogle Scholar
- Hosch W, Stiller W, Mueller D, Gitsioudis G, Welzel J, Dadrich M, Buss SJ, Giannitsis E, Kauczor HU, Katus HA, Korosoglou G: Reduction of radiation exposure and improvement of image quality with BMI-adapted prospective cardiac computed tomography and iterative reconstruction. Eur J Radiol. 2012, 81: 3568-3576. 10.1016/j.ejrad.2011.06.055.View ArticlePubMedGoogle Scholar
- Funama Y, Taguchi K, Utsunomiya D, Oda S, Yanaga Y, Yamashita Y, Awai K: Combination of a low-tube-voltage technique with hybrid iterative reconstruction (iDose) algorithm at coronary computed tomographic angiography. J Comput Assist Tomogr. 2011, 35: 480-485. 10.1097/RCT.0b013e31821fee94.PubMed CentralView ArticlePubMedGoogle Scholar
- Schoenhagen P, Bolen MA, Halliburton SS: Iterative CT reconstruction of aortic intramural hematoma. Circ J. 2011, 75: 1774-1776. 10.1253/circj.CJ-10-1250.View ArticlePubMedGoogle Scholar
- Noel PB, Fingerle AA, Renger B, Rummeny EJ, Doherty C: Clinical comparison study of a novel statistical iterative and filtered backprojection reconstruction. Medical Imaging 2011: Physics of Medical Imaging. Proceedings of SPIE 7961. Edited by: Pelc NJ, Samei E, Nishikawa RM. 2011Google Scholar
- Philips Healthcare: iDose4 Iterative Reconstruction Technique. Breakthrough in image Quality and Dose Reduction With the 4th Generation of Reconstruction. 2011, Eindhoven: Philips HealthcareGoogle Scholar
- Maxfield MW, Schuster KM, McGillicuddy EA, Young CJ, Ghita M, Bokhari SA, Oliva IB, Brink JA, Davis KA: Impact of adaptive statistical iterative reconstruction on radiation dose in evaluation of trauma patients. J Trauma Acute Care Surg. 2012, 73: 1406-1411. 10.1097/TA.0b013e318270d2fb.PubMed CentralView ArticlePubMedGoogle Scholar
- Deutsche Gesellschaft für Unfallchirurgie: S3-Leitlinie "Polytrauma/Schwerverletzten-Behandlung" (AWMF Register-Nr. 012/019). DGU Mitteil Nachr. 2011, 33: 1-188.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.