- Open Access
Emergency cricothyrotomy – a systematic review
Scandinavian Journal of Trauma, Resuscitation and Emergency Medicinevolume 21, Article number: 43 (2013)
An emergency cricothyrotomy is the last-resort in most airway management protocols and is performed when it is not possible to intubate or ventilate a patient. This situation can rapidly prove fatal, making it important to identify the best method to establish a secure airway. We conducted a systematic review to identify whether there exists superiority between available commercial kits versus traditional surgical and needle techniques.
Medline, EMBASE and other databases were searched for pertinent studies. The inclusion criteria included manikin, animal and human studies and there were no restrictions regarding the professional background of the person performing the procedure.
In total, 1,405 unique references were identified; 108 full text articles were retrieved; and 24 studies were included in the review. Studies comparing kits with one another or with various surgical and needle techniques were identified. The outcome measures included in this systematic review were success rate and time consumption. The investigators performing the studies had chosen unique combinations of starting and stopping points for time measurements, making comparisons between studies difficult and leading to many conflicting results. No single method was shown to be better than the others, but the size of the studies makes it impossible to draw firm conclusions.
The large majority of the studies were too small to demonstrate statistically significant differences, and the limited available evidence was of low or very low quality. That none of the techniques in these studies demonstrated better results than the others does not necessarily indicate that each is equally good, and these conclusions will likely change as new evidence becomes available.
Emergency cricothyrotomy has long been established as the last-resort and potentially life-saving procedure for patients who cannot be intubated or ventilated and would otherwise face imminent death. It is essential to identify the situation and perform an emergency cricothyrotomy before the patient ends up with a cardiac arrest [1–3]. How often “cannot intubate, cannot ventilate” (CICV) situations followed by the need for a surgical airway occur depends on the location of the patient, the qualifications and experience of the health care provider, and the medical condition of the patient. The published incidence numbers of CICV situations with the use of surgical airway techniques vary from 0 to 18.5% in the studies included in Table 1[4–19]. However, these figures should be read with caution since some figures are old and may not be representative for the present practice in the relevant services. Furthermore, the incidence will probably vary depending on how early in the applied airway management algorithm it is recommended to perform an emergency cricothyrotomy.
Traditionally, a surgical airway has been established by making an incision through the skin and the cricothyroid membrane into the tracheal lumen through which an endotracheal tube is inserted . This method is not without complications, and several investigators have modified the technique [21, 22]. In recent years, a number of commercial kits that include all of the necessary equipment to establish a surgical airway have reached the market. These commercial kits can be divided into two broad categories. One category depends on puncture of the cricothyroid membrane with a needle, followed by the insertion of a guidewire through the needle (Seldinger technique) . A specially designed endotracheal tube included in the kit can be inserted into the trachea over the guidewire, if necessary, after the prior use of a dilator. The other category of commercial kits does not rely on the use of a guidewire; in these kits, a cutting device is employed to create a lumen in the cricothyroid membrane that is wide enough to accommodate the endotracheal tube included in the kit. Furthermore, some authors recommend that a temporary airway be established by puncturing the cricothyroid membrane with a wide-bore cannula through which the patient can be ventilated and oxygenated .
Because the CICV-situation is encountered infrequently, most health care providers have little if any clinical experience with the various techniques that can be employed. It is unclear which of the available methods that is most reliable and fastest to perform in the extremely time-critical CICV situations. The goal of this systematic review was to identify the current available research literature to evaluate the evidence-based information on this topic.
The electronic databases that were searched included the following: MEDLINE, EMBASE, CINAHL, Cochrane Central Register of Controlled Trials (CENTRAL), and British Nursing Index.
The full search strategies are available in the Appendix. The last update of the searches was performed on April 10, 2013. No language restriction was enforced.
The bibliographies of the included studies were also examined for other studies that could be considered for inclusion.
Types of studies
Cricothyrotomy is an intervention that is infrequently performed; when performed, it constitutes an emergency procedure. For ethical and practical reasons, we expected few, if any, randomised clinical studies to have been conducted, as well as that most studies – independent of design – would most likely include few patients. Based upon these considerations, all prehospital, hospital and laboratory studies with living patients/participants, cadavers, manikins and animals were included.
All study designs were included.
Types of participants
Cricothyrotomy may be necessary for patients of all ages suffering from either medical conditions or trauma. Hence, no limitations were enforced based on the age or condition (or trauma) the patient suffered.
All categories of health care providers, independent of formal training and education, could end up in a situation where a cricothyrotomy seems like a necessary intervention. Hence, no limitations were implemented in the literature searches on the background of the personnel participating in the reported studies.
Types of interventions
The three Seldinger kits included in this review are the Arndt emergency cricothyrotomy catheter set (Arnd) (Cook, Bloomington, IL, USA), the Melker emergency cricothyrotomy (Mlkr) (Cook, Bloomington, IL, USA) and the Minitrach II (Mini) (Smiths Medical Ltd, Hythe, UK) (Table 2). Common to all kits is the performance of an initial skin incision with a scalpel before the cricothyroid membrane is punctured with a needle that is attached to a syringe. When aspiration of air confirms the tracheal position of the needle, a guidewire is inserted through the needle. Following the removal of the needle, a specially designed endotracheal tube is introduced into the tracheal lumen over the guidewire, and the guidewire is removed. An inflatable airway bag is connected to the tube, and the patient can be ventilated.
The Airfree coniotomy set (Airf) (FRC Medizintechnik, Holzheim a.F., Deutschland), the Patil’s airway (Pati) (Cook, Bloomington, IL, USA), the Portex cricothyrotomy kit (PCK) (Smiths Medical Ltd, Hythe, UK), the QuickTrach1 kit (QT1) and the QuickTrach2 kit (QT2) (VBM Medizintechnik GmbH, Sulz, Germany), the TracheoQuick emergency coniotomy set (Trqu) (Teleflex Medical GmbH, Kernen, Deutschland), and the Pertrach kit (Pert) (Pulmodyne, Indianapolis, IN, USA) are the seven kits that do not rely on the Seldinger technique that have been identified in this review. In these kits, custom-made cutting devices are used to incise the cricothyroid membrane, and no guidewire is used. After the dilation of the hole, a specially designed endotracheal tube can be introduced into the tracheal lumen. Again, an inflatable airway bag is connected to the tube, and the patient can be ventilated.
Airf consists of a tube surrounding a sharp trocar. The trocar is used to incise the skin and the cricothyroid membrane, and when placed in the tracheal lumen, the trocar is withdrawn while the surrounding tube remains in place, allowing the ventilation of the patient. The QT1 and QT2 are similar to Airf, and both methods involve an artificial airway pre-loaded over a large bore needle and a direct puncture of the cricothyroid membrane. When the correct position is confirmed by aspiration of air, the needle is removed. The QT1 and QT2 can be connected to the ventilation bag with the provided flexible tubing. The Pati and Trqu are based upon the same principle, as is the Pert, but with the Pert, a splitting needle is employed.
The PCK is based on a tube-over-needle design through which the correct placement of the spring-loaded needle in the trachea is shown by a flag in the needle hub indicating tissue contact. Once the tracheal lumen has been reached, the indicator flag in the needle hub disappears, reappearing when the needle touches the posterior tracheal wall. After redirecting and advancing the device 1-2 cm caudally, the needle is removed and the cricothyrotomy tube is slid over the dilator into the tracheal lumen, and finally, the dilator is removed.
There are a number of variants of the standard surgical technique (Surg) described in the literature. Most variants make use of a scalpel, a dilator, hemostats, a tracheal hook and a tracheostomy tube . The operator will make a skin incision over the cricothyroid membrane. The membrane is localised by blunt dissection before a short horizontal stab incision is made in the lower part of the membrane. The larynx is stabilised with the tracheal hook at the inferior aspect of the thyroid cartilage, and the incision in the membrane is dilated before the tracheostomy tube is advanced into the tracheal lumen.
An alternative surgical technique makes use of the Bair claw device (Bair), which can be attached to a scalpel . A horizontal incision is made through the cricothyroid membrane after the palpation of the landmarks. The scalpel is withdrawn from the airway, and the device is rotated caudally before the hooks are spread with blunt dissection of the tissue. Finally, the endotracheal tube is inserted between the hooks of the device.
The rapid four-step technique (RFST) was developed from the classic surgical emergency cricothyrotomy . The cricothyroid membrane is palpated before a horizontal stab incision is made through the skin and membrane with the scalpel, followed by tracheal hook traction in the caudal direction. This allows the operator to perform the procedure with minimal assistance, holding the tracheal hook with one hand while passing the endotracheal tube with the other hand. The bougie-assisted cricothyrotomy technique (BACT) is a refinement of the RFST . After the incision through the skin and the cricothyroid membrane, a bougie is used to secure the tracheal lumen before the endotracheal tube is advanced into the trachea over the bougie.
The cricothyrotomy scissor (Csci) is a modified pair of scissors that are pushed with closed blades without previous incision of the skin all the way through the cricothyroid membrane into the trachea . Inside the trachea, the scissors are opened, and the hole is enlarged. After closing the blades, the Csci are rotated 90 degrees; the blades are again opened; and an endotracheal tube can be passed through the resulting hole.
A needle cricothyrotomy (Need) is performed with a cannula attached to a syringe . The needle is advanced through the skin and underlying tissues until the cricothyroid membrane is punctured. Aspiration of air confirms the correct intratracheal placement. The cannula is then advanced over the needle until the flanges rest on the skin and the needle is removed. An alternative to employing a wide-bore cannula intended for intravenous use is the Cook Transtracheal Jet Airway Catheter (Trac) (Cook, Bloomington, IL, USA), which can be combined with a jet ventilator. Once the tracheal catheter placement is achieved, ventilation is initiated using the manual jet ventilator connected to a high-flow oxygen source.
We have included all identified studies where two or more of the above techniques have been compared.
Types of outcome measures
The two outcome measures studied in this review were success rate and time used to secure the airway. Most of the identified studies also had other outcome measures like complication rate and preferred technique of the performer, but success rate and time consumption were the outcome measures that were always reported.
MS assessed all references at the title/abstract level, while PKH, ARN and SL each independently assessed a third of the references. Disagreements were resolved through discussion between the two assessors, and when required, one of the other authors was consulted. We obtained full text articles of all studies that were not discarded on the abstract level.
Data extraction and management
We designed a form to extract data. For eligible studies, two review authors independently extracted the information. Discrepancies were resolved through discussion.
Assessment of the risk of bias in the included studies
Two review authors independently assessed the risk of bias for each study using the criteria outlined in the Cochrane Handbook for Systematic Reviews of Interventions . Disagreements were resolved by discussion or through the involvement of a third assessor.
The risk of bias tool used for randomised controlled trials involves assessing the following five criteria:
Sequence generation (checking for possible selection bias)
Allocation concealment (checking for possible selection bias)
Blinding (checking for possible performance bias and detection bias)
Incomplete outcome data (checking for possible attrition bias through withdrawals, dropouts, protocol deviations, and use of ITT analyses where appropriate)
Selective reporting bias (checking if expected outcomes are reported and if there is reason to suspect publication bias)
Measures of treatment effect
For success rate, the results are presented as summary risk ratios (RR) with 95% confidence intervals (CI).
The time consumption has been presented in descriptive tables with median and IQR if mentioned in the original paper. The time consumption for the procedure when the procedure failed (secure airways not obtained) was handled differently in different studies. Some studies presented the time consumption from successful placements only, excluding the failures. Other studies used a stop rule where if more than a set number of seconds were used, they were classified as failures; in these, the stop rule number of seconds were presented as the time consumption.
Analysis and synthesis
Where we considered it appropriate to combine results from different studies, we have done so. Where we considered it inappropriate, we presented the results descriptively in tables. We carried out statistical analysis (meta-analyses) using the RevMan 5 software (RevMan 2011, http://ims.cochrane.org/revman). We expected that there would be differences among trials in both the populations and interventions, so we used random effects meta-analysis for combining data.
Assessment of heterogeneity
The size and direction of the effects have been considered and consulted with the I2 and Chi-square statistics to quantify the level of heterogeneity among the trials in each analysis. Caution in the interpretation of the results is advised where substantial (I2 between 30 and 60%) or considerable (I2 between 50 and 100%) heterogeneity exists.
Grading the quality of the evidence
The quality of the evidence for each of the critically important outcomes has been graded using the GRADE methodology (http://www.gradeworkinggroup.org). For each outcome, the quality of the evidence was assessed using the eight GRADE criteria: five considering downgrading, including study limitations, heterogeneity, directness of the evidence, precision, and reporting bias, and three considering possible upgrading, including strong effect, dose-response, and plausible confounding.
The systematic literature searches yielded 1,405 unique references, including a meta-analysis performed by Hubble and coworkers . A total of 108 full text articles were retrieved, and two authors read them independently; 24 studies were included in this review (Figure 1) [21, 22, 24, 25, 29–48]. The 85 studies that were excluded in this process did not include sufficient information on comparison between two or more techniques to be included in this review. All the remaining 24 studies were prospective experimental studies with varying degrees of randomisation (Table 3). Studies involving human cadavers (ten studies), various airway simulators (eight studies), a pig laryngeal model (three studies), anaesthetised sheep (two studies) and sheep cadavers (one study) were included. The interventions were performed by students and professionals from a variety of disciplines (anaesthesiology, emergency medicine, intensive care unit physicians, medical students, paramedics). In the study involving the largest number of participants, 64 anaesthesiologists took part, while the smallest study in this respect involved two anaesthesiologists. Studies were included from the following eight countries: USA (eight studies), Germany (five studies), Austria (three studies), Australia (two studies), Ireland (two studies), United Kingdom (two studies) and one each from Canada and The Netherlands. The oldest study was published in 1993, and the most recent was published in 2012.
The risk of bias of the included studies is summarised in Figure 2. Methods of randomisation and allocation were poorly described in the majority of the articles. Therefore, we have concluded with an unclear risk of bias for most of the studies.
Nine groups of comparisons were included:
Comparison of two kits based upon the Seldinger guidewire technique
Arnd, Mlkr and Mini are all based upon the Seldinger guidewire technique, but only one study was identified in which two techniques based upon this principle were compared . In this study comparing Mlkr and Mini, it was found that Mini had a significantly higher success rate than Mlkr.
Comparison of one kit based upon the Seldinger guidewire technique with a kit that is not based upon this technique
The Mlkr kit has been compared to Pati, PCK, QTI and QT2, respectively, and Mini has been compared with QT1 [25, 30–32, 34–37, 40, 42]. For all comparisons, no significant difference in the success rates between the two devices was detected.
Comparison of one kit based upon the Seldinger guidewire technique with an open surgical technique
Both Arnd, Mlkr and Mini have been compared to Surg [25, 32, 34, 36, 38, 39, 41, 45],. In one study, Mlkr has been compared with Csci . For all comparisons, no significant difference in the success rates between the two devices was detected. In one study, Arnd was found to be a statistically faster technique than Surg. In five of the seven studies comparing Mlkr and Surg, it was shown that Surg is statistically faster. Csci was found to be significantly faster than Mlkr in the single study comparing the two devices.
Comparison of one kit based upon the Seldinger guidewire technique with a needle technique
Comparison of two kits not based upon the Seldinger guidewire technique
We have identified the following comparisons between two such kits: Airf vs. PCK, QT1, QT2 and Trqu, respectively; PCK vs. QT1, QT2 and Trqu, respectively; QT1 vs. QT2 and Trqu, respectively; as well as Pati vs. QT1 and QT2 vs. Trqu [29, 32, 42]. No significant difference in success rate was observed in any of the comparisons. In one of the studies comparing PCK and QT2, it was reported that QT2 was significantly faster than PCK.
Comparison of one kit not based upon the Seldinger guidewire technique with an open, surgical technique
A variety of the surgical technique has been compared to Airf, Pert, PCK, QT1, QT2 and Trqu [25, 29, 32–34, 36, 47, 48]. In one study, Csci and QT1 has been compared . For none of the comparisons, a significant difference in success was found. In the studies where Airf and Trqu, respectively, were compared to Surg, it was found that Surg was a significantly slower way to establish a surgical airway than the alternatives. The same result was found in one of the three studies where QT1 and Surg were compared. In contrast, in the single study comparing Pert and Surg, Surg was found to be the faster technique. The same result was achieved in one of the studies comparing PCK and Surg.
Comparison of one kit not based upon the Seldinger guidewire technique with a needle technique
Comparison of two open, surgical techniques
Varieties of Surg have been compared to Bair, RFST and Csci, respectively [21, 24, 25, 44]. Furthermore, in one study RFST and BACT have been compared . No significant difference in success rate has been reported for these comparisons. In all studies involving Surg, the alternative (that is Bair, RFST and Csci, respectively) was found to be a significantly faster option. In the single study comparing RFST and BACT, the difference in time consumption was statistically significant showing that BACT was the faster technique.
Comparison of one open, surgical technique with a needle technique
This is to our knowledge the first systematic review comparing all commercial kits designed to perform emergency cricothyrotomy with surgical and needle techniques. The main result of this review is that no technique has been proven to be superior to the others, regarding success rate or time consumption. The quality of evidence is low or very low for several reasons. The studies are uniformly small, so that even though a number of studies comparing two or more techniques have been published, relatively few events have actually been analysed. When evidence is drawn from small studies, the results are uncertain and normally contain large confidence intervals. A common consequence of small trials is heterogeneity among studies, of which there are multiple examples in this review. A tendency exists toward many unique comparisons, rather than more general comparisons, and there is large variation among the roles of health care providers who perform these procedures. However, there were no studies involving surgeons. This can be interpreted that the perceived success rate with surgical techniques for experienced surgeons is so high that they do not find the commercial kits an interesting alternative even though there are no studies supporting this notion.
The studies were performed on a number of models varying from human cadavers and dead animal models to a multitude of airway models. It was difficult to make direct comparisons among studies since the primary study authors to a large degree had defined their own unique starting and stopping points in the time measurements.
Emergency cricothyrotomies are performed under stressful conditions and severe time pressure. If unsuccessful, these procedures can prove fatal or severely disabling for patients. In a laboratory setting, it is very demanding to achieve the same level of stress. Furthermore, it can be necessary to perform emergency cricothyrotomies under suboptimal conditions, such as prehospital settings, in which a lack of light, background noise and entrapped patients may add to the difficulty. Of the studies identified in this review, only one study included these types of factors that most likely affect both the success rate and the time consumption [28, 30]. Furthermore, as a result of the models used, the procedures were performed without bleeding. In real life situations, bleeding will occur, thus increasing the level of difficulty of the procedure.
In the majority of the studies, there were no upper time limit that should not be exceeded for the procedure to be accepted as successful, and – with one exception – in the studies that operated with an upper limit it was between 180 and 300 seconds. An emergency cricothyrotomy will usually be performed in CICV-situation when other devices have failed and the clinical condition of the patient has already started to deteriorate. In such situations, the operator does not have 180 to 300 seconds at their disposal to perform the procedure. Salah and coworkers on the other hand chose to publish the success rate after 40 seconds, an – in our opinion – clinically much more realistic time frame, and none of the participating anaesthesiologists succeeded to establish a secure airway with Mini in this study even after five attempts . In contrast, after five attempts approximately two thirds of the participants managed to establish a secure airway with the surgical technique. This study underscore the point that focusing on success rate only is insufficient; time aspects are also essential. In general, the difference in success rates may reflect varying definition of success, and operator experience, but study model probably also has an influence on outcome.
Arnd, Mlkr and Mini are the three Seldinger-based techniques included in this review, and we only identified a single study where two Seldinger-based techniques were compared. The study in which Mlkr and Mini were compared was the only study where a statistically significant difference in success rate between two devices/techniques was found. Mini was significantly faster than Mlkr, and in this study, Mini had a 100% success rate in contrast to the 74% success rate of Mlkr.
In the five different comparisons between a kit based upon the Seldinger technique and a kit not based upon this technique (Mlkr vs. Pati, Mlkr vs. PCK, Mlkr vs. QT1, Mlkr vs. QT2 and Mini vs. QT1), no statistically significant difference in the success rate was found. However, these studies were small, as were all the studies included in the review, making it necessary to interpret these results with great caution. The success rate is not the only important parameter in the context of emergency cricothyrotomies; time consumption is also critical. Individual studies tended to use their own unique combinations of starting and stopping points, making the evaluation of time comparisons difficult. Even so, it is evident that in some cases, the Seldinger-based technique was faster than the non-Seldinger technique, while in other studies, the opposite result was found. This could partly be due to different studies using different techniques and time measurements. However, when Mlkr and PCK were compared, for instance, two studies found that Mlkr was the faster technique, while two studies concluded that PCK was the faster technique. One might suspect that in those studies where anaesthesiologists performed the procedures, the Seldinger-based techniques would be faster because of the familiarity anaesthesiologists have with this technique. However, in the three comparisons of Mlkr and PCK performed by anaesthesiologists, PCK was faster than Mlkr in two of the three studies.
In four studies, a Seldinger-based technique was compared with a surgical technique (Arnd vs. Surg, Mlkr vs. Surg, Mlkr vs. Csci, Mini vs. Surg), and no statistical difference in success rate was detected. In the large majority of the comparisons, the surgical technique was faster than the Seldinger technique, even though in one study, the opposite result was found. These studies indicate that surgical techniques are generally faster than Seldinger-based kits, while both techniques have high success rates. This may indicate that surgical techniques should be favored by most healthcare providers. However, it is not unreasonable to assume that the lack of bleeding in the majority of the study models may – especially for the surgical techniques – result in artificially short procedure times and high success rates .
In the two studies where a Seldinger-based technique was compared with a needle technique (Mlkr vs. Trac, Mini vs. Need), the success rate was 100% for all procedures. Regarding time consumption, the results conflicted, with one study showing that the Seldinger-based technique was faster and the other showing the opposite result. This discrepancy may be explained by the fact that neither the Seldinger-based technique nor the needle technique used was the same in the two studies.
Seven different comparisons between a non-Seldinger kit and an open surgical technique were found (Airf vs. Surg, Pert vs. Surg., PCK vs. Surg, QT1 vs. Surg, QT2 vs. Surg, QT1 vs. Csci, Trqu vs. Surg). It is a weakness that five of the seven comparisons were only performed once, but this is the case for most of the currently published comparisons in this field. Conflicting results were obtained. In some instances, the non-Seldinger technique was faster, while in other instances the surgical technique was. Even in the two comparisons that were studied in more than one publication, the results conflicted, ultimately not indicating which technique was faster. The success rates were uniformly high for all devices in all studies.
There were also seven different comparisons of a non-Seldinger kit compared to a needle technique (Airf. vs. Need, Pati vs. Trac, PCK vs. Need, QT1 vs. Need, QT2 vs. Need, QT2 vs. Trac, Trqu vs. Need). All comparisons were performed only once. Each study was small, and no significant differences in success rates were detected. With the exception of the comparison between QT2 and Trac, the needle technique was found to be the faster technique, although the time differences were small and insignificant. Furthermore, it has been argued that needle techniques – in contrast to the other techniques - do not result in an airway that leads to adequate oxygenation and ventilation due to the small calibre of the artificial airway. The needle may kink and is not suitable for patient transport.
Four different comparisons between two surgical techniques were identified (Surg vs. Bair, Surg vs. RFST, Surg vs. Csci, RFST vs. BACT). All techniques had a high success rate, and no significant difference in success rate was found. However, these studies were also small, and, as was the case with all comparisons performed, any difference must be pronounced to have been detected. However, significant time differences were shown to indicate that BACT was faster than RFST, which in turn was faster than Surg. Csci was also significantly faster than Surg, but this technique has never been compared to RFST or BACT.
The field of techniques for establishing an emergency cricothyrotomy is hampered by the many very small studies performed. There is limited evidence of low and very low quality comparing these different emergency techniques for use in CICV situations. That none of the techniques produced better results than the others in these studies does not necessarily indicate that they are all equally good, and these conclusions will likely change as new evidence becomes available. This review does not justify recommending one technique over others. Success may rely on the operator’s experience and skill/training and not on the technique chosen.
Ovid MEDLINE(R) In-Process & Other Non-Indexed Citations and Ovid MEDLINE(R) 1946 to Present
Search strategy: 10.04.2013
Cricoid Cartilage/su [Surgery] 879
Airway Obstruction/su, th [Surgery, Therapy] 4493
exp Airway Management/ 84568
Neck/su [Surgery] 2893
((difficult or definite or management or surgery or surgical or obstruction* or control) adj2 airway*).tw. 19262
(airway adj5 emergenc*).tw. 981
((can?t or "can not" or cannot or diffucult* or inabilit*) adj2 (ventilat* or intubate*)).tw. 296
((cricoid or neck) and (surgery or emergency or emergencies)).tw. 23509
(Otolaryngology or otorhinolaryngology).tw. 9401
(single adj stab).tw. 28
(transtracheal adj2 ventilat*).tw. 203
(jet adj2 ventilat*).tw. 1276
exp Emergency Treatment/ 88882
(emergency or emergencies or surgery or surgical).tw. 1258060
(18 or 19) and (20 or 21 or 22) 3661
17 or 23 141620
(cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*).tw. 1342
24 and 25 678
((quicktrach or seldinger or portex or pck or minitrach or melker) adj3 (device* or kit or technique* or set* or tube*)).tw. 774
(portex or cook or vbm).ti,ab. 4656
((difficult or definite or management or surgery or surgical or obstruction* or control) adj2 airway*).tw. 19262
(cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*).tw. 1342
27 or 28 5305
29 or 30 20326
31 and 32 100
33 not 26 67
Embase 1980 to 2013 Week 14
Search strategy: 10.04.2013
cricoid/su [Surgery] 255
airway obstruction/su, th [Surgery, Therapy] 3455
exp assisted ventilation/ 96166
((difficult or definite or management or surgery or surgical or obstruction* or control) adj2 airway*).tw. 23818
((can?t or "can not" or cannot or diffucult* or inabilit*) adj2 (ventilat* or intubate*)).tw. 403
(airway adj5 (emergency or emergencies)).tw. 1176
((cricoid or neck) and (surgery or emergency or emergencies)).tw. 33129
(otorhinolaryngology or otolaryngology).tw. 13719
(single adj stab).tw. 30
(transtracheal adj2 ventilat*).tw. 223
(jet adj2 ventilat*).tw. 1580
emergency/ or Emergency treatment/ 45263
Surgery/ or ear nose throat surgery/ or emergency surgery/ or "head and neck surgery"/ 207679
(emergency or emergencies or surgery or surgical).tw. 1555446
23 and 24 3656
17 or 25 169870
(cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*).tw. 1571
26 and 27 783
(quicktrach or seldinger or portex or pck or minitrach or melker).ti,ab,dm,dv. 2593
(portex or cook or vbm).ti,ab,dm,dv. 11677
29 or 30 13274
((difficult or definite or management or surgery or surgical or obstruction* or control) adj2 airway*).tw. 23818
(cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*).tw. 1571
32 or 33 25016
31 and 34 344
35 not 28 273
Clinical Trials: 34
Economic Evaluations: 2
#1 (cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*) (39)
#1 (quicktrach or seldinger or portex or pck or minitrach or melker) 184
#2 (portex or cook or vbm) 2997
#3 ((difficult or definite or management or surgery or surgical or obstruction* or control) NEAR/2 airway*) 2123
#4 (cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*) 39
#5 (( #1 OR #2 ) AND ( #3 OR #4 )) 49
#6 (#5 AND NOT #4) 45
TI ( cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro* ) or AB ( cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro* )
(MH "Cricothyrotomy") (87)
S32 S30 and S31 (25)
S31 S28 or S29 (1102)
S30 S25 or S26 or S27 (6159)
S29 portex or cook or vbm (925)
S28 TX quicktrach or seldinger or portex or pck or minitrach or melker (227)
S27 TX difficult N2 airway or definite N2 airway or management N2 airway or surgery N2 airway or surgical N2 airway or obstruction* N2 airway or control N2 airway (5976)
S26 TI ( cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro* ) or AB ( cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro* ) (198)
S25 (MH "Cricothyrotomy") (72)
Hits: 69 + 27
#2 Search #1 Limits: published in the last 1 year
#1 Search cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*
#58 Search #54 AND (#55 or #56) Limits: published in the last 1 year
#57 Search #54 AND (#55 or #56)
#56 Search cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*
#55 Search (difficult or definite or management or surgery or surgical or obstruction* or control) and airway
#54 Search ((quicktrach or seldinger or portex or pck or minitrach or melker) ) OR portex or cook or vbm
British Nursing Index
(cricothyr* or cricotomy or cricothracheotomy or thyrocrico* or intercricothyro*).tw
Airfree coniotomy set
Arndt emergency cricothyrotomy catheter set
Bougie-assisted cricothyrotomy technique
Bair claw device
Cannot intubate, cannot ventilate
Melker emergency cricothyrotomy set
Portex cricothyrotomy kit
QuickTrach 2 kit
Rapid four-step technique
Standard surgical technique
Transtracheal jet airway catheter
TracheoQuick emergency coniotomy set.
Berlac P, Hyldmo PK, Kongstad P, Kurola J, Nakstad AR, Sandberg M: Pre-hospital airway management: guidelines from a task force from the Scandinavian Society for Anaesthesiology and Intensive Care Medicine. Acta Anaesthesiol Scand. 2008, 52 (7): 897-907. 10.1111/j.1399-6576.2008.01673.x.
Henderson JJ, Popat MT, Latto IP, Pearce AC, Society DA: Difficult Airway Society guidelines for management of the unanticipated difficult intubation. Anaesthesia. 2004, 59 (7): 675-694. 10.1111/j.1365-2044.2004.03831.x.
Apfelbaum JL, Hagberg CA, Caplan RA, Blitt CD, Connis RT, Nickinovich DG, Hagberg CA, Caplan RA, Benumof JL, Berry FA: Practice guidelines for management of the difficult airway: an updated report by the american society of anesthesiologists task force on management of the difficult airway. Anesthesiology. 2013, 118 (2): 251-270. 10.1097/ALN.0b013e31827773b2.
Adams BD, Cuniowski PA, Muck A, De Lorenzo RA: Registry of emergency airways arriving at combat hospitals. J Trauma. 2008, 64 (6): 1548-1554. 10.1097/TA.0b013e3181728c41.
Germann CA, Baumann MR, Kendall KM, Strout TD, McGraw K: Performance of endotracheal intubation and rescue techniques by emergency services personnel in an air medical service. Prehosp Emerg Care. 2009, 13 (1): 44-49. 10.1080/10903120802474505.
Stephens CT, Kahntroff S, Dutton RP: The success of emergency endotracheal intubation in trauma patients: a 10-year experience at a major adult trauma referral center. Anesth Analg. 2009, 109 (3): 866-872. 10.1213/ane.0b013e3181ad87b0.
Sagarin MJ, Barton ED, Chng Y-M, Walls RM, National Emergency Airway Registry Investigators: Airway management by US and Canadian emergency medicine residents: a multicenter analysis of more than 6,000 endotracheal intubation attempts. Ann Emerg Med. 2005, 46 (4): 328-336. 10.1016/j.annemergmed.2005.01.009.
Adnet F, Jouriles NJ, Le Toumelin P, Hennequin B, Taillandier C, Rayeh F, Couvreur J, Nougière B, Nadiras P, Ladka A: Survey of out-of-hospital emergency intubations in the French prehospital medical system: a multicenter study. Ann Emerg Med. 1998, 32 (4): 454-460. 10.1016/S0196-0644(98)70175-1.
McIntosh SE, Swanson ER, Barton ED: Cricothyrotomy in air medical transport. J Trauma. 2008, 64 (6): 1543-1547. 10.1097/TA.0b013e3181271b60.
Bair AE, Panacek EA, Wisner DH, Bales R, Sakles JC: Cricothyrotomy: a 5-year experience at one institution. J Emerg Med. 2003, 24 (2): 151-156. 10.1016/S0736-4679(02)00715-1.
Sakles JC, Laurin EG, Rantapaa AA, Panacek EA: Airway management in the emergency department: a one-year study of 610 tracheal intubations. Ann Emerg Med. 1998, 31 (3): 325-332. 10.1016/S0196-0644(98)70342-7.
Fortune JB, Judkins DG, Scanzaroli D, McLeod KB, Johnson SB: Efficacy of prehospital surgical cricothyrotomy in trauma patients. J Trauma. 1997, 42 (5): 832-836. 10.1097/00005373-199705000-00013.
Jacobson LE, Gomez GA, Sobieray RJ, Rodman GH, Solotkin KC, Misinski ME: Surgical cricothyroidotomy in trauma patients: analysis of its use by paramedics in the field. J Trauma. 1996, 41 (1): 15-20. 10.1097/00005373-199607000-00004.
Nugent WL, Rhee KJ, Wisner DH: Can nurses perform surgical cricothyrotomy with acceptable success and complication rates?. Ann Emerg Med. 1991, 20 (4): 367-370. 10.1016/S0196-0644(05)81656-7.
Robinson KJ, Katz R, Jacobs LM: A 12-year experience with prehospital cricothyrotomies. Air Med J. 2001, 20 (6): 27-30. 10.1016/S1067-991X(01)70006-5.
Boyle MF, Hatton D, Sheets C: Surgical cricothyrotomy performed by air ambulance flight nurses: a 5-year experience. J Emerg Med. 1993, 11 (1): 41-45. 10.1016/0736-4679(93)90008-U.
McGill J, Clinton JE, Ruiz E: Cricothyrotomy in the emergency department. Ann Emerg Med. 1982, 11 (7): 361-364. 10.1016/S0196-0644(82)80362-4.
Brown J, Thomas F: What happens with failed blind nasal tracheal intubations?. Air Med J. 2001, 20 (2): 13-16. 10.1016/S1067-991X(01)70093-4.
Cook TM, Woodall N, Frerk C, Fourth National Audit P: Major complications of airway management in the UK: results of the Fourth National Audit Project of the Royal College of Anaesthetists and the Difficult Airway Society. Part 1: anaesthesia. Br J Anaesth. 2011, 106 (5): 617-631. 10.1093/bja/aer058.
American College of Surgeons, Committee on Trauma: Cricothyroidotomy. Advanced Trauma Life Support. 2012, 58-61.
Holmes JF, Panacek EA, Sakles JC, Brofeldt BT: Comparison of 2 cricothyrotomy techniques: standard method versus rapid 4-step technique. Ann Emerg Med. 1998, 32 (4): 442-446. 10.1016/S0196-0644(98)70173-8.
Hill C, Reardon R, Joing S, Falvey D, Miner J: Cricothyrotomy technique using gum elastic bougie is faster than standard technique: a study of emergency medicine residents and medical students in an animal lab. Acad Emerg Med. 2010, 17 (6): 666-669. 10.1111/j.1553-2712.2010.00753.x.
Seldinger SI: Catheter replacement of the needle in percutaneous arteriography; a new technique. Acta Radiol. 1953, 39 (5): 368-376. 10.3109/00016925309136722.
Bair AE, Sakles JC: A comparison of a novel cricothyrotomy device with a standard surgical cricothyrotomy technique. Acad Emerg Med. 1999, 6 (11): 1172-1174. 10.1111/j.1553-2712.1999.tb00123.x.
Schober P, Hegemann M, Schwarte L, Loer S, Noetges P: Emergency cricothyrotomy-A comparative study of different techniques in human cadavers. Resuscitation. 2009, 80 (2): 204-209. 10.1016/j.resuscitation.2008.10.023.
Higgins JPT, Altman DG: Assessing risk of bias in included studies. Cochrane handbook for systematic reviews of interventions. Edited by: Higgins JPT, Green S. 2008, Wiley, 187-235.
Guyatt G, Oxman AD, Akl EA, Kunz R, Vist G, Brozek J, Norris S, Falck-Ytter Y, Glasziou P, Debeer H: GRADE guidelines: 1. Introduction—GRADE evidence profiles and summary of findings tables. J Clin Epidemiol. 2011, 64 (4): 383-394. 10.1016/j.jclinepi.2010.04.026.
Hubble MW, Wilfong DA, Brown LH, Hertelendy A, Benner RW: A meta-analysis of prehospital airway control techniques part II: alternative airway devices and cricothyrotomy success rates. Prehosp Emerg Care. 2010, 14 (4): 515-530. 10.3109/10903127.2010.497903.
Helmstaedter V, Wetsch WA, Bottiger BW, Hinkelbein J: [Comparison of ready-to-use devices for emergency cricothyrotomy : randomized and controlled feasibility study on a mannequin]. Anaesthesist. 2012, 61 (4): 310-319. 10.1007/s00101-012-2008-2.
Givens GC, Shelton SL, Brown EA: Emergency cricothyrotomy in confined space airway emergencies: a comparison. Prehospital Disaster Med. 2011, 26 (4): 259-261. 10.1017/S1049023X11006352.
Metterlein T, Frommer M, Ginzkey C, Becher J, Schuster F, Roewer N, Kranke P: A randomized trial comparing two cuffed emergency cricothyrotomy devices using a wire-guided and a catheter-over-needle technique. J Emerg Med. 2011, 41 (3): 326-332. 10.1016/j.jemermed.2010.04.008.
Murphy C, Rooney SJ, Maharaj CH, Laffey JG, Harte BH: Comparison of three cuffed emergency percutaneous cricothyroidotomy devices to conventional surgical cricothyroidotomy in a porcine model. Br J Anaesth. 2011, 106 (1): 57-64. 10.1093/bja/aeq294.
Salah N, Mhuircheartaigh RN, Hayes N, McCaul C: A comparison of four techniques of emergency transcricoid oxygenation in a manikin. Anesth Analg. 2010, 110 (4): 1083-1085.
Mariappa V, Stachowski E, Balik M, Clark P, Nayyar V: Cricothyroidotomy: comparison of three different techniques on a porcine airway. Anaesth Intensive Care. 2009, 37 (6): 961-967.
Benkhadra M, Lenfant F, Nemetz W, Anderhuber F, Feigl G, Fasel J: A comparison of two emergency cricothyroidotomy kits in human cadavers. Anesth Analg. 2008, 106 (1): 182-185. 10.1213/01.ane.0000296457.55791.34.
Dimitriadis JC, Paoloni R: Emergency cricothyroidotomy: a randomised crossover study of four methods. Anaesthesia. 2008, 63 (11): 1204-1208. 10.1111/j.1365-2044.2008.05631.x.
Assmann NM, Wong DT, Morales E: A comparison of a new indicator-guided with a conventional wire-guided percutaneous cricothyroidotomy device in mannequins. Anesth Analg. 2007, 105 (1): 148-154. 10.1213/01.ane.0000266439.43797.9e.
Sulaiman L, Tighe SQM, Nelson RA: Surgical vs wire-guided cricothyroidotomy: a randomised crossover study of cuffed and uncuffed tracheal tube insertion. Anaesthesia. 2006, 61 (6): 565-570. 10.1111/j.1365-2044.2006.04621.x.
Schaumann N, Lorenz V, Schellongowski P, Staudinger T, Locker GJ, Burgmann H, Pikula B, Hofbauer R, Schuster E, Frass M: Evaluation of Seldinger technique emergency cricothyroidotomy versus standard surgical cricothyroidotomy in 200 cadavers. Anesthesiology. 2005, 102 (1): 7-11. 10.1097/00000542-200501000-00005.
Fikkers BG, van Vugt S, van der Hoeven JG, van den Hoogen FJA, Marres HAM: Emergency cricothyrotomy: a randomised crossover trial comparing the wire-guided and catheter-over-needle techniques. Anaesthesia. 2004, 59 (10): 1008-1011. 10.1111/j.1365-2044.2004.03794.x.
Keane MF, Brinsfield KH, Dyer KS, Roy S, White D: A laboratory comparison of emergency percutaneous and surgical cricothyrotomy by prehospital personnel. Prehosp Emerg Care. 2004, 8 (4): 424-426.
Vadodaria BS, Gandhi SD, McIndoe AK: Comparison of four different emergency airway access equipment sets on a human patient simulator. Anaesthesia. 2004, 59 (1): 73-79. 10.1111/j.1365-2044.2004.03456.x.
Mutzbauer TS, Munz R, Helm M, Lampl LA, Herrmann M: [Emergency cricothyrotomy–puncture or anatomical preparation? Peculiarities of two methods for emergency airway access demonstrated in a cadaver model]. Anaesthesist. 2003, 52 (4): 304-310. 10.1007/s00101-003-0476-0.
Davis DP, Bramwell KJ, Hamilton RS, Chan TC, Vilke GM: Safety and efficacy of the rapid rour-step technique for cricothyrotomy using a Bair claw. J Emerg Med. 2000, 19 (2): 125-129. 10.1016/S0736-4679(00)00197-9.
Eisenburger P, Laczika K, List M, Wilfing A, Losert H, Hofbauer R, Burgmann H, Bankl H, Pikula B, Benumof JL: Comparison of conventional surgical versus Seldinger technique emergency cricothyrotomy performed by inexperienced clinicians. Anesthesiology. 2000, 92 (3): 687-690. 10.1097/00000542-200003000-00012.
Chan TC, Vilke GM, Bramwell KJ, Davis DP, Hamilton RS, Rosen P: Comparison of wire-guided cricothyrotomy versus standard surgical cricothyrotomy technique. J Emerg Med. 1999, 17 (6): 957-962. 10.1016/S0736-4679(99)00123-7.
Johnson DR, Dunlap A, McFeeley P, Gaffney J, Busick B: Cricothyrotomy performed by prehospital personnel: a comparison of two techniques in a human cadaver model. Am J Emerg Med. 1993, 11 (3): 207-209. 10.1016/0735-6757(93)90125-U.
Helm M, Hossfeld B, Jost C, Lampl L, Bockers T: Emergency cricothyroidotomy performed by inexperienced clinicians--surgical technique versus indicator-guided puncture technique. Emerg Med J. 2012, Epub ahead of print
Paix BR, Griggs WM: Emergency surgical cricothyroidotomy: 24 successful cases leading to a simple ‘scalpel-finger-tube’ method. Emerg Med Australas. 2012, 24 (1): 23-30. 10.1111/j.1742-6723.2011.01510.x.
All authors declare that they have no competing interest regarding the devices and the techniques presented in this review.
All authors participated in all phases of the process of writing this systematic review. All authors read and approved the final manuscript.
Electronic supplementary material
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.