- Original research
- Open Access
The impact of environmental factors in pre-hospital thermistor-based tympanic temperature measurement: a pilot field study
© Skaiaa et al. 2015
Received: 3 February 2015
Accepted: 8 September 2015
Published: 24 September 2015
Few pre-hospital services have the possibility to accurately measure core temperature (T core ). Non-invasive estimation of T core will improve pre-hospital decision-making regarding the triage and management of hypothermic patients. Thermistor-based tympanic temperature (T tymp ) correlates well with T core in controlled studies; however, little is known about the feasibility of using T tymp under field conditions. This study assessed the impact of pre-hospital environmental factors on the accuracy of T tymp . Deep rectal temperature (T rect ) was used as a substitute for T core .
Normothermic volunteers (n = 13) were exposed to four simulated field conditions producing local cooling of the head and ear canal. After exposure, T tymp was recorded every 15 s for 10 min and compared with T rect . Descriptive analysis and Bland-Altman plots were used to assess agreement.
Immediately after exposure mean T tymp was low, but increased rapidly and reached an apparent steady state after 3–5 min. After 5 and 10 min, the mean temperature difference (∆T rect-tymp ) ranged from 1.5–3.2 °C (SD = 0.5) and 1.2–2.0 °C, respectively. T rect remained unchanged throughout the study period.
After surface cooling of head and neck, T tymp did not accurately reflect core temperature within the first 10 min of measurement. The variation of ∆T rect-tymp was low after 10 min, regardless of the initial degree of cooling. With the risk of over-triage, T tymp may at this point provide an indication of T core and also exhibit a trend.
A core temperature (T core ) < 30 °C prior to cerebral hypoxia (i.e. avalanche, drowning) or out-of-hospital-cardiac-arrest (OHCA) is associated with increased chances of survival and a neurological favourable outcome [1–8]. For most other patients, accidental hypothermia is associated with increased morbidity and mortality and should be treated aggressively [9–16]. Early measurement of T core is important to diagnose and treat hypothermia and will, for OHCA, aid the decision to either continue resuscitation to hospital or declare the patient dead on-scene [17, 18]. Temperature measurement becomes increasingly useful in rural and wilderness areas with long transports and in mass casualty incidents when resources are limited. Unfortunately, T core is infrequently evaluated in the field, and many pre-hospital services do not have adequate equipment to accurately diagnose hypothermia [10, 19, 20]. Ideally, T core is measured in the distal third of the oesophagus [21, 22], but requires the patient to be sedated or intubated. Common non-invasive techniques include the ear canal, nasopharynx and rectum. Historically, one of the most accepted techniques for field use is the tympanic thermistor-based temperature (T tymp ). Studies and case reports have shown that T tymp accurately reflects brain temperature; therefore, it has been accepted as a good non-invasive estimation of T core [23–25]. Several factors may influence the accuracy of T tymp , and T tymp has been shown to decrease in cold ambient temperatures . Other mechanisms that might influence accuracy include wind and direct cooling of the external auditory canal, as well as equipment failure in cold or wet conditions. The aim of this study was to investigate the impact of environmental factors on the accuracy of T tymp under field conditions.
In this observational study we compared T tymp with deep rectal temperature (T rect ) in healthy and normothermic volunteers during four separate exposure scenarios designed to simulate probable real-life situations. Because T rect accurately reflects T core in normothermic patients in a stable environment [22, 27], it was defined as T core in this study.
To ensure normothermia, the volunteers were outdoors only during the scenarios. They were placed sitting on a 14-mm sleeping mat (Mammut Bamse Extreme, Mammut Sports Group, Switzerland) in a shady area to avoid direct sunlight. All volunteers were dressed similarly: in a base layer of woollen thermal underwear, with an outer shell of an expedition down jacket with a hood and Gore-Tex® pants. A woollen hat and mittens were used for insulation of head and hands. The lower body was covered with a thermal rescue bag (Fjellduken Thermo Hunter, Jerven AS, Norway).
Each volunteer served as his or her own control. Each exposure scenario was analysed as an entity to ensure as similar ambient conditions as possible, but also to allow a minimum of 100 min of indoor rest between each scenario to avoid carry-over effects.
T rect was measured using a standard multi-purpose probe (Mon-a-Therm, General purpose Temperature Probe 12Fr/Ch, Covidien, Ireland) attached to an external monitor (Propaq Encore 206EL, ZOLL Medical Corporation, MA, USA). The probe was placed 20 cm into the rectum and taped to the skin.
A thermistor-based thermometer (Métraux®, Crissier, Switzerland) was used to measure T tymp . The thermometers accuracy was verified at Oslo University Hospital, Dep. of Medical Technology. In the temperature range of 22.4–38.0 °C it was found to display an accuracy of ± 0.2 °C (OUS Rikshospitalet, MTV, ref.no. 942252, Oct. 2013). The sensor consists of two thin and malleable plastic-covered metal wires integrated into a soft rubber earplug. After each exposure scenario the ear sensor was inserted into the external auditory canal. The earplug was securely taped to the external ear to avoid dislocation during the tests. We were able to visually read T tymp in real time from the instrument’s display. Otoscopy was performed before the first insertion and after the completion of all tests.
Ambient temperature was measured with an external monitor (Propaq Encore 206EL, ZOLL Medical Corporation, MA, USA). Wind speed was measured using a Silva wind gauge instrument (Silva ADC Wind, Silva Sweden AB, Sweden).
Ambient air without local insulation/Wind (Ins -). Simulation of a normal outdoor setting. The volunteer was exposed to ambient air and wind without protection of the head for 10 min before placement of the ear sensor. No additional insulation of the head/ear was applied during data collection.
Ambient air with local insulation/Wind (Ins +). Exposure identical to scenario A., but after placement of the ear sensor the head was covered with a woollen hat and the down jacket hood was engaged.
Snow in ear canal/Snow (Ins +). Simulation of a victim buried in an avalanche. Snow was manually packed densely into the external auditory canal and left for 4 min. After 2 min, any melted snow was replaced with fresh snow. After 4 min, the remaining snow was removed, and the external auditory canal was dried for 3–4 s with a towel before placement of the ear sensor. Finally, the head was covered with a woollen hat and the down jacket hood was engaged.
Icy water in ear canal/Water (Ins +). Simulation of a victim immersed in cold water. The volunteer submerged his/her entire head and neck in a 12-liter bucket of icy water (1-4 °C) for 5–10 s. Submersion was conducted three consecutive times, with a 1-min rest between submersion periods. Immediately after the final submersion, the external auditory canal was dried for 3–4 s with a towel before placement of the ear sensor. Finally, the head was covered with a woollen hat and the down jacket hood was engaged.
Inclusion required that volunteers be older than 18 years of age, non-smokers, and healthy individuals. Volunteers were excluded if otoscopy revealed pathology in the external auditory canal.
The study took place at 860 m above sea level during 2 days in May 2014. T rect and T tymp baseline was initially measured after a minimum of 30 min at rest indoors in a median ambient temperature of 20.5 °C (19.5–21.1). The volunteers were then moved outdoors in random pairs. T rect was measured before each scenario (baseline) and then every 15 s during the first 20 scenarios. As T rect remained unchanged (≤0.1 °C), we then reduced the measuring interval to every 10 min (before, during and after each scenario). T tymp was recorded every 15 s for 10 min after each exposure scenario. T tymp baseline was not repeated before each scenario to avoid pre-warming of the sensor. Ambient temperature and wind speed were recorded every 30 min throughout both study days.
We used descriptive analysis (median, minimum, maximum, percentiles, and confidence intervals) and graphical methods to characterize the temperature trajectories. The comparison of the measurement methods was done with Bland-Altman plots with 95 % limits of agreement. P-values less than 0.05 were considered significant. All computation and graphics were completed using Matlab 7.10 (Mathworks Inc).
This study was approved by the Regional Ethics Committee (REK sør-øst, ref 2014/108).
A total of 13 volunteers (nine women) were included. Written informed consent was obtained. Median age was 25 years (range 20–39), median BMI 22.6 kg/m2 (range 20.2–27.5). Median T rect at baseline was 37.4 °C (range 36.8–37.6). The variation of T rect between baseline, before, during and after each scenario was ≤ 0.1 °C and hence considered constant. Median T tymp at baseline was 36.5 °C (range 36.0–37.5) and significantly lower than T rect (P < 0.001). Respectively, the median ambient temperature and wind speed during both study days were 1.6 °C (range 0.5–5.0) and 5.5 m/s (range 2.7–10.0).
The first obtained T tymp was very low compared to T rect for all exposure scenarios. In correlation with previous studies , T tymp then increased rapidly to an apparent steady state within a few minutes. After 5 and 10 min, mean ∆T rect-tymp varied up to 3.2 and 2.0 °C, respectively. The thermometer used in this study did not accurately estimate the core body temperature, neither at baseline nor throughout the study. Rather than providing a true value, T tymp gave an increasingly stable indication of T core , as confirmed in the Bland-Altman plots. The almost constant T rect , leading to Bland-Altman plots dominated by T tymp , caused the linear trend observed in the plots. This effect increased with the variation of T tymp , i.e. it was clearest visible directly after cooling and weakened with the reduction of the variation of T tymp over time. The extremely low concordance correlations may be caused by the low variation in T rect as well, since all correlation type analysis depend on the range of the considered variables.
Non-invasive thermometry will normally be utilized when the patient is not intubated or in the absence of invasive T core devices. T tymp will in these situations give an increasingly stable indication of T core , with the risk of modest over-triage. Drying the ear canal thoroughly and leaving the ear sensor in place for a minimum of 10 min will reduce this risk. In addition, in a field situation one can expect T core to decrease further during on-site management and transport (with the possible exception of a well-circulated patient with intact shivering capacity that is re-warmed en-route). This reduces the significance of an initial over-triage for especially patients in moderate (T core 28–32 °C) and deep (T core <28 °C) hypothermia. The low variation after 10 min indicates that measurement of T tymp may enable the health provider to monitor the T core trend during treatment and transport.
Insulation of the head and neck with a woollen hat combined with a thick down jacket hood did not significantly improve accuracy. Possible explanations for this include cold equipment or insufficient ear padding, which could allow for heat loss through convection and evaporation. More likely, the ambient temperature might not have been cold enough to reveal a true benefit.
When exposed to rapid temperature changes, T rect can lag behind T core . To avoid such changes, and to ensure normothermia, we insulated the participants well and reduced outdoor exposure times to a minimum. Deep T rect remained stable during the scenarios and was therefore considered an accurate estimation of T core in this study.
We did not repeat baseline T tymp before exposure, as this would imply warming of the thermometer just prior to the post-exposure data collection and create a carry-over effect. As T rect demonstrated stable core temperatures throughout the study we hypothesized that also T tymp baseline would not change significantly after 100 min of indoor rest. However, a systematic bias cannot be excluded.
Despite standardization, a field study may be biased by changes in temperature and wind. After the volunteers were exposed to snow and icy water, all visible snow and water was removed from the ear before placing the temperature probe; however, the amount of residual snow or water cannot be estimated. Each of the exposure scenarios reflected a real-life situation. Although we did not measure the time used to attach and fix the thermometer, in most cases the first reading was less than 20 s after the cooling had stopped. Time is often a limiting factor in the clinical setting, and time spent to obtain access to a suitable temperature measurement site is reportedly one of several factors that contribute to low compliance in the field .
Clinical implications and conclusions
In the field, a thermistor-based T tymp will initially display very low values, with an apparent steady state to be expected after 3–5 min. T tymp will however continue to increase after 10 min, and may at this point overestimate the degree of hypothermia up to a median of 2.0 °C in an ear canal cooled from snow or water. The tympanic probe should therefore be left in place in a dry ear canal for more than 10 min before estimating T core as a basis for decision-making. The variation of T tymp after 5 and 10 min was low in this study, indicating that T tymp, in a non-OHCA setting, may be used with reasonable accuracy to exhibit the temperature trend during management and transport.
Future research should focus on developing non-invasive thermometer devices designed for rapid placement, better insulation against the environment, and improved accuracy.
The authors wish to thank all the volunteers who participated in the tests. The Norwegian Air Ambulance Foundation (SNLA), the Division of Emergencies and Critical Care, Oslo University Hospital Rikshospitalet, the Department of Anaesthesia and Intensive care, Haukeland University Hospital, Bergen, and the Department of Clinical Engineering Haukeland University Hospital, Bergen, are thanked for financial support and for providing medical equipment and help.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Tipton MJ, Golden FSC. A proposed decision-making guide for the search, rescue and resuscitation of submersion (head under) victims based on expert opinion. Resuscitation. 2011;82(7):819–24. doi:10.1016/j.resuscitation.2011.02.021.View ArticlePubMedGoogle Scholar
- Brugger H, Durrer B, Elsensohn F, Paal P, Strapazzon G, Winterberger E, et al. Resuscitation of avalanche victims: Evidence-based guidelines of the international commission for mountain emergency medicine (ICAR MEDCOM): intended for physicians and other advanced life support personnel. Resuscitation. 2013;84(5):539–46. doi:10.1016/j.resuscitation.2012.10.020.View ArticlePubMedGoogle Scholar
- Farstad M, Andersen KS, Koller ME, Grong K, Segadal L, Husby P. Rewarming from accidental hypothermia by extracorporeal circulation. A retrospective study. Eur J Cardiothorac Surg. 2001;20(1):58–64.View ArticlePubMedGoogle Scholar
- Wanscher M, Agersnap L, Ravn J, Yndgaard S, Nielsen JF, Danielsen ER, et al. Outcome of accidental hypothermia with or without circulatory arrest: experience from the Danish Præstø Fjord boating accident. Resuscitation. 2012;83(9):1078–84. doi:10.1016/j.resuscitation.2012.05.009.View ArticlePubMedGoogle Scholar
- Boue Y, Lavolaine J, Bouzat P, Matraxia S, Chavanon O, Payen JF. Neurologic recovery from profound accidental hypothermia after 5 hours of cardiopulmonary resuscitation. Crit Care Med. 2014;42(2):e167–70. doi:10.1097/CCM.0b013e3182a643bc.View ArticlePubMedGoogle Scholar
- Hilmo J, Naesheim T, Gilbert M. “Nobody is dead until warm and dead”: Prolonged resuscitation is warranted in arrested hypothermic victims also in remote areas - A retrospective study from northern Norway. Resuscitation. 2014;85(9):1204–11.View ArticlePubMedGoogle Scholar
- Pasquier M, Zurron N, Weith B, Turini P, Dami F, Carron P-N, et al. Deep Accidental Hypothermia with Core Temperature Below 24°C Presenting with Vital Signs. High Alt Med Biol. 2014;15(1):58–63.View ArticlePubMedGoogle Scholar
- Sawamoto K, Bird SB, Katayama Y, Maekawa K, Uemura S, Tanno K, et al. Outcome from severe accidental hypothermia with cardiac arrest resuscitated with extracorporeal cardiopulmonary resuscitation. Am J Emerg Med. 2014;32(4):320–4. doi:10.1016/j.ajem.2013.12.023.View ArticlePubMedGoogle Scholar
- Danzl DF, Pozos RS, Auerbach PS, Glazer S, Goetz W, Johnson E, et al. Multicenter hypothermia survey. Ann Emerg Med. 1987;16(9):1042–55.View ArticlePubMedGoogle Scholar
- Brandstrom H, Johansson G, Giesbrecht GG, Angquist KA, Haney MF. Accidental cold-related injury leading to hospitalization in northern Sweden: an eight-year retrospective analysis. Scand J Trauma Resusc Emerg Med. 2014;22:6. doi:10.1186/1757-7241-22-6.PubMed CentralView ArticlePubMedGoogle Scholar
- Rajagopalan S, Mascha E, Na J, Sessler DI. The effects of mild perioperative hypothermia on blood loss and transfusion requirement. Anesthesiology. 2008;108(1):71–7.View ArticlePubMedGoogle Scholar
- Reynolds L, Beckmann J, Kurz A. Perioperative complications of hypothermia. Best Pract Res Clin Anaesthesiol. 2008;22(4):645–57.View ArticlePubMedGoogle Scholar
- Gentilello LM, Jurkovich GJ, Stark MS, Hassantash SA, O'Keefe GE. Is hypothermia in the victim of major trauma protective or harmful? A randomized, prospective study. Ann Surg. 1997;226:439–47. doi:10.1097/00000658-199710000-00005. discussion 47–49.PubMed CentralView ArticlePubMedGoogle Scholar
- Ireland S, Endacott R, Cameron P, Fitzgerald M, Paul E. The incidence and significance of accidental hypothermia in major trauma-A prospective observational study. Resuscitation. 2011;82(3):300–6.View ArticlePubMedGoogle Scholar
- Lapostolle F, Sebbah J, Couvreur J, Koch F, Savary D, Tazarourte K, et al. Risk factors for onset of hypothermia in trauma victims: The HypoTraum study. Crit Care. 2012;16(4):R142–R.PubMed CentralView ArticlePubMedGoogle Scholar
- Søreide K. Clinical and translational aspects of hypothermia in major trauma patients: from pathophysiology to prevention, prognosis and potential preservation. Injury. 2014;45(4):647–54. doi:10.1016/j.injury.2012.12.027.View ArticlePubMedGoogle Scholar
- Soar J, Perkins GD, Abbas G, Alfonzo A, Barelli A, Bierens JJLM, et al. European Resuscitation Council Guidelines for Resuscitation 2010 Section 8. Cardiac arrest in special circumstances: Electrolyte abnormalities, poisoning, drowning, accidental hypothermia, hyperthermia, asthma, anaphylaxis, cardiac surgery, trauma, pregna. Resuscitation. 2010;81(10):1400–33. doi:10.1016/j.resuscitation.2010.08.015.View ArticlePubMedGoogle Scholar
- Brown DJ, Brugger H, Boyd J, Paal P. Accidental hypothermia. N Engl J Med. 2012;367(20):1930–8. doi:10.1056/NEJMra1114208.View ArticlePubMedGoogle Scholar
- Harten-Ash LV, Hudson A. Hypothermia among trauma patients in the Emergency Department (ED): a review of documentation and management. Scand J Trauma Resusc Emerg Med. 2014;22 Suppl 1:10. doi:10.1186/1757-7241-22-s1-p10.View ArticleGoogle Scholar
- Karlsen AM, Thomassen O, Vikenes BH, Brattebø G. Equipment to prevent, diagnose, and treat hypothermia: a survey of Norwegian pre-hospital services. Scand J Trauma Resusc Emerg Med. 2013;21:63. doi:10.1186/1757-7241-21-63.PubMed CentralView ArticlePubMedGoogle Scholar
- Robinson J, Charlton J, Seal R, Spady D, Joffres MR. Oesophageal, rectal, axillary, tympanic and pulmonary artery temperatures during cardiac surgery. Can J Anaesth. 1998;45(4):317–23. doi:10.1007/BF03012021.View ArticlePubMedGoogle Scholar
- Lefrant JY, Muller L, de La Coussaye JE, Benbabaali M, Lebris C, Zeitoun N, et al. Temperature measurement in intensive care patients: comparison of urinary bladder, oesophageal, rectal, axillary, and inguinal methods versus pulmonary artery core method. Intensive Care Med. 2003;29(3):414–8.PubMedGoogle Scholar
- Walpoth BH, Galdikas J, Leupi F, Muehlemann W, Schlaepfer P, Althaus U. Assessment of hypothermia with a new “tympanic” thermometer. J Clin Monit. 1994;10(2):91–6.View ArticlePubMedGoogle Scholar
- Sato KT, Kane NL, Soos G, Gisolfi CV, Kondo N, Sato K. Reexamination of tympanic membrane temperature as a core temperature. J Appl Physiol. 1996;80(4):1233–9.PubMedGoogle Scholar
- Oberhammer R, Beikircher W, Hörmann C, Lorenz I, Pycha R, Adler-Kastner L, et al. Full recovery of an avalanche victim with profound hypothermia and prolonged cardiac arrest treated by extracorporeal re-warming. Resuscitation. 2008;76(3):474–80.View ArticlePubMedGoogle Scholar
- Strapazzon G, Procter E, Avancini G, Uberbacher N, Putzer G, Hofer G, et al. Thermistor probe for measurement of tympanic temperature: Influence of ambient temperature. Resuscitation. 2013;84:S93–S. doi:10.1016/j.resuscitation.2013.08.236.View ArticleGoogle Scholar
- Cork RC, Vaughan RW, Humphrey LS. Precision and accuracy of intraoperative temperature monitoring. Anesth Analg. 1983;62(2):211–4.PubMedGoogle Scholar