- Original research
- Open Access
Quality of cardiopulmonary resuscitation in out-of-hospital cardiac arrest before and after introduction of a mechanical chest compression device, LUCAS-2; a prospective, observational study
- Tinne Tranberg1Email author,
- Jens F Lassen†1,
- Anne K Kaltoft†1,
- Troels M Hansen†2, 3, 4,
- Carsten Stengaard†1,
- Lars Knudsen†4,
- Sven Trautner†5 and
- Christian J Terkelsen†1
© Tranberg et al.; licensee BioMed Central. 2015
- Received: 12 January 2015
- Accepted: 9 April 2015
- Published: 22 April 2015
Mechanical chest compressions have been proposed to provide high-quality cardiopulmonary resuscitation (CPR), but despite the growing use of mechanical chest compression devices, only few studies have addressed their impact on CPR quality. This study aims to evaluate mechanical chest compressions provided by LUCAS-2 (Lund University Cardiac Assist System) compared with manual chest compression in a cohort of out-of-hospital cardiac arrest (OHCA) cases.
In this prospective study conducted in the Central Denmark Region, Denmark, the emergency medical service attempted resuscitation and reported data on 696 non-traumatic OHCA patients between April 2011 and February 2013. Of these, 155 were treated with LUCAS CPR after an episode with manual CPR. The CPR quality was evaluated using transthoracic impedance measurements collected from the LIFEPAK 12 defibrillator, and the effect was assessed in terms of chest compression rate, no-flow time and no-flow fraction; the fraction of time during resuscitation in which the patient is without spontaneous circulation receiving no chest compression.
The median total episode duration was 21 minutes, and the episode with LUCAS CPR was significantly longer than the manual CPR episode, 13 minutes vs. 5 minutes, p < 0.001. The no-flow fraction was significantly lower during LUCAS CPR (16%) than during manual CPR (35%); difference 19% (95% CI: 16% to 21%; p < 0.001). No differences were found in pre- and post-shock no-flow time throughout manual CPR and LUCAS CPR.
Contrary to the manual CPR, the average compression rate during LUCAS CPR was in conformity with the current Guidelines for Resuscitation, 102/minute vs. 124/minute, p < 0.001.
Mechanical chest compressions provided by the LUCAS device improve CPR quality by significantly reducing the NFF and by improving the quality of chest compression compared with manual CPR during OHCA resuscitation. However, data on end-tidal Co2 and chest compression depth surrogate parameters of CPR quality could not be reported.
- Out-of-hospital cardiac arrest (OHCA)
- CPR quality
- Mechanical chest compression
High-quality chest compression and early defibrillation are particularly essential for survival outcome after out-of-hospital cardiac arrest (OHCA) . Recent guidelines for resuscitation underline the importance of quality chest compressions and of minimising the time with no chest compressions during cardiopulmonary resuscitation (CPR) . However, studies have shown that chest compressions performed by health care professionals do not meet the recommendations for compression rate, depth and continuity, which results in considerably longer no-flow times than necessary and desirable [3,4].
Despite the growing use of these devices, randomised studies in humans have not been able to show better outcome for OHCA patients resuscitated with mechanical chest compressions [7-9]. Research comparing the effect of manual CPR and mechanical CPR in the same patient is limited. The present study aimed to implement LUCAS in the physician-manned prehospital critical care teams and the helicopter emergency medical service (HEMS) and to evaluate if the mechanical chest compressions provided by LUCAS improve CPR quality compared with manual chest compressions in OHCA patients.
The emergency medical service (EMS) in the Central Denmark Region is organised as a two-tier system; it operates a double-dispatch service that covers an area of 13,142 km2 inhabited by a total of 1.3 million people. The first tier consists of 75 conventional ambulances manned with two EMS providers. The second tier consists of nine physician-manned prehospital critical care teams and one HEMS available 24/7. The prehospital critical care teams and the HEMS have the competency to provide advanced life support (ALS). ALS includes the potential use of the LUCAS device. Conventional ambulances provide basic life support (BLS) and defibrillation only.
The EMS in the Central Region Denmark has a standardised pre-hospital-resuscitation protocol, which was strictly adhered to in this study. A conventional ambulance is dispatched to all emergencies including OHCA. It normally arrives at the scene as a first responder. The prehospital critical care team or the HEMS is dispatched to patients with presumed OHCA as determined by the dispatcher triage and according to availability. All cardiac arrest patients are treated according to the 2010 Guidelines for Resuscitation. During the entire study period, the physicians of the prehospital critical care team and the HEMS serve jointly as the decision-maker who decides whether or not the LUCAS device is to be used in the attempt of resuscitation. In patients receiving LUCAS CPR, the EMS providers initiate the BLS until the arrival of the physician-manned mobile prehospital critical care team bringing the LUCAS device. Manual CPR is continued when the LUCAS device is being deployed; and in order to minimise the interruptions in CPR, it is only paused briefly when the back plate is inserted and the arms positioned. When the prehospital critical care team or the HEMS are involved, the patient is transported directly to the tertiary university hospital’s Heart Centre, which is staffed with dedicated anaesthesiologists, cardiologists and surgeons. Provided that active treatment is indicated, acute coronary angiography (CAG) and/or percutaneous coronary intervention (PCI) is performed, and therapeutic hypothermia is applied according to international guidelines. The patients are subsequently treated in the Cardiac Intensive Care Unit which is staffed by anaesthesiologists and cardiologists. However, if the prehospital critical care team or the HEMS are unable to assist the EMS providers, the operating procedure for the EMS providers state that after three defibrillation attempts or in the case of a non-shockable rhythm after three 2-minute cycles of CPR and persisting cardiac arrest, the patient is transported to the nearest hospital.
A total of nine LUCAS devices were introduced in the prehospital critical care team and the HEMS. Prior to this introduction, the physicians and the EMS providers were informed about the study protocol and the procedures, and they were instructed how to handle the LUCAS device. Furthermore, they accomplished a manikin-scenario-training-session, which enabled them to use the device correctly, i.e. to deploy the device within 20 seconds while minimising interruptions in CPR. The physicians and the EMS providers who did not have the opportunity to attend the training session were trained and accredited by one of the physicians who took part in the session.
The inclusion criteria for the present study were non-traumatic OHCA, age above 18 years and OHCA occurring between 1 October 2011 and 31 January 2013. Exclusion criteria were pregnancy, trauma, intoxication, inability to attach the LUCAS device to the patient, patients with missing data and total resuscitation episode < 2 minutes.
Standard LIFEPAK 12 defibrillators (Physio-Control, Redmond, WA, USA) were used. De defibrillators continuously measure the transthoracic impedance (TTI) by applying a near constant current across the defibrillations pads. After a CPR effort, the ECGs and the TTI data of the OHCA case were transferred to a central server at the tertiary hospital; Aarhus University Hospital, Denmark. Furthermore, the prehospital critical care teams and the HEMS filled out a separate study form regarding the end-tidal CO2 (ETCO2), which served as a surrogate marker of CPR quality.
The software programme CODE-STAT™ -8 (Physio-Control Inc., Redmond, WA) was used to process the TTI data (Figure 1B, C, D). The software automatically annotates the chest compressions; however, each OHCA case was verified at the level of single compressions. Any incorrect automatic annotations were deleted, new annotations were added when annotations were missing; and the system calculated pre-shock and post-shock pauses, the compression rate and the actual number of compressions per minute, no-flow time (NFT) and no-flow fraction (NFF). The NFT, defined as the time without return of spontaneous circulation (ROSC) and the time without chest compressions, is a validated measure of the CPR quality, and it is reported according to previously published definitions [11-13]. The term NFF is defined as NFF = (NFT/episode duration – time with ROSC). The NFF represents the proportion of interruptions in CPR during the episode. The actual number of chest compressions delivered per minute represents both the compression rate and the pauses in the compressions.
We defined the CPR-pause interval as the time from the trailing edge of the last chest compression to the leading edge of the next chest compression. Mechanical compressions were distinguished from the manual compressions by their highly regular morphology (Figure 1C). Usable data files included TTI data for the entire episode, from the first manual chest compression to the end of the final mechanical chest compression. The case was excluded if the total episode was shorter than 2 minutes.
Additionally, nationally adapted Utstein style forms filled out by the EMS personnel were used to obtain data on bystander CPR, OHCA location and the use of an automated external defibrillator (AED) prior to arrival of the EMS and whether OHCA was witnessed or unwitnessed. Data on the patients who achieved ROSC, survived to hospital discharge and survived after 30 days were collected from the OHCA registration form, ambulance records and hospital records.
The properties of LUCAS-2
LUCAS is a chest compression system that provides both active compression and decompression of the chest wall back to neutral position (Figure 1A). The first generation of the device, LUCAS-1, was driven by compressed air which was superseded by a battery-driven device in 2009. The device consists of a silicone rubber suction cup that is applied to the patient’s chest and a cylinder mounted on two legs connected to a stiff back palate. It delivers compressions at a rate of 102 per minute and at a depth between 5 and 6 cm as prescribed in the guidelines for resuscitation provided by The European Resuscitation Council (ERC).
Data management and statistical calculations were performed using STATA/SE 12.1. Normally distributed continuous variables are presented as mean ± standard deviation and non-normally distributed variables as median and quartiles. Differences were analysed with Student’s paired t-test or Wilcoxon’s sign rank test as appropriate. Categorical variables are presented as number (n) and percent (%) as appropriate, and differences are analysed with McNemar’s test. The a priori α-level was 0.05.
Data were reported on 696 OHCA patients who were attempted resuscitated between 1 October 2011 and 31 January 2013. Of these, 196 were resuscitated with both manual CPR and LUCAS CPR. Due to incomplete TTI data, 31 (4%) patients were excluded. Another 41 (21%) patients were excluded due to the fact that only data with LUCAS CPR were available. The remaining 155 OHCA patients comprised the study population (Figure 2).
Baseline characteristics, n = 155
Cardiac arrest background variables
Age in years, mean (SD)
66 (±15), n = 146
Male gender 67%
Place of cardiac arrest
Cardiac arrest witnessed
Health care professional
Rhythm on arrival of EMS
Proportion of AED analysis
Defibrillation with AED before arrival of the EMS
Among the patients with a shockable rhythm as the presenting heart rhythm, the time (IQR) from emergency call to first defibrillation by the EMS was 9 (8–19) minutes.
Quality of cardiopulmonary resuscitation (CPR)
n = 155
n = 155
Episode duration, min.
5 (2; 6)
13 (11; 14)
No-flow fraction, %
35 (33; 37)
16 (15; 18)
Chest compression rate per minute
124 (121; 126)
102 (102; 102)
Number of chest compressions per minute
75 (72; 79)
84 (82; 85)
No-flow time (NFT) and rhythm analysis with/without defibrillation
Mean (95% CI) sec.
Number with data
Mean (95% CI) sec.
Number with data
NFT in relation to AED analysis
15 (13; 18)
16 (13; 19)
NFT in relation to AED analysis
17 (16; 18)
18 (16; 20)
17 (15; 20)
20 (16; 23)
7 (6; 8)
7 (6; 9)
There was no significant difference between the results of either NFT during rhythm analysis with and without defibrillation or pre- and post-shock NFT with manual or LUCAS CPR (Table 3). LUCAS NFT was median (IQR = 24 (14–38)) seconds.
In-hospital treatment among patients admitted alive (n = 45) and among patients admitted with on-going LUCAS CPR (n = 7), n = 52
Admitted alive (n = 45)
Ongoing LUCAS CPR (n = 7)
Total (n = 52)
Percutaneous coronary intervention
While recent years have seen studies reporting on the outcome associated with mechanical CPR [7,14], only few studies have provided data on the quality of mechanical chest compressions, and those that have are limited by a low number of patients . The present study evaluates the performance and quality of both manual and mechanical CPR in the same patient primarily based on TTI.
Our results show that during the manual episode, NFF lasted 34% of the time in patients without ROSC. However, once the LUCAS device was deployed, the NFF was reduced significantly to 16% of the time. Although an NFF of 34% during manual CPR is lower than what has been reported in previous studies where CPR was performed according to the former resuscitation guidelines of 2005 , we still find that the NFF is too high. Its size stresses the importance of short pauses during resuscitation and the importance of offering CPR training with performance feedback to further improve CPR quality. The average NFF of only 16% achieved with the LUCAS device was low. One could speculate that rhythm analysis was not performed every second minute as prescribed by the current guidelines. However, we have no data to substantiate this because the CODE-STAT does not allow registrations of the manual rhythm analysis. Thus, ambulance personnel have to use the automatic defibrillator mode as documented in the data derived from the defibrillator; inversely, physicians normally interpret the rhythm themselves to reduce CPR interruptions.
The low NFF with the LUCAS device may also have been achieved owing to fewer interruptions while loading the patient into the ambulance and during transport with on-going LUCAS.
Another advantage of the LUCAS device is that it affords the possibility of delivering shocks during compression and enables users to shorten or eliminate pauses for defibrillation. Nevertheless, we found no difference in pre-shock and post-shock NFT in relation to automated rhythm analysis throughout the time with manual CPR or LUCAS CPR. The median pre-shock pause was 17 sec. (IQR 15–20) and the median post-shock pause was 7 sec. (IQR 6–8). While these pauses are reasonable for defibrillation in automatic mode during manual CPR and similar to pauses reported in previous research , they are suboptimal for defibrillation during LUCAS CPR. However, the TTI analysis of all cases at the level of single chest compression revealed that the pre-shock and the post-shock NFT during LUCAS CPR might be overestimated due to the inability of CODE-STAT to register chest compressions when shock is delivered. The chest compression rate and the actual number of compressions per minute, which are important elements in CPR quality, were significantly better performed during LUCAS CPR. The chest compressions were delivered at a rate higher than recommended, and they were probably also too shallow during manual CPR (124 compressions per minute), which may result in failure to achieve ROSC.
Although they fell short of being optimal, the manual chest compression rate and the number of chest compressions per minute still appear superior to the manual compression rates (138 compressions per minute) reported in the study by Krarup et al. .
The episode duration with LUCAS CPR was nearly three times longer than the manual CPR episode (Table 3). This agrees well with previous reports, and it presumably reflects that these cases are the ones with the longest resuscitation attempt .
In a study of changes in bystander resuscitation attempts and survival during a 10-year period in which national initiatives were taken to improve rates of bystander CPR, Wissenberg et al.  concluded that an increase in survival following OHCA was associated with an increase in bystander CPR, regardless of witnessed cardiac arrest. This is consistent with the findings in our study, which showed a large proportion of patients with cardiac arrest at home and a high rate of bystander CPR. Our study was not designed or statistically powered to evaluate survival outcome. Nevertheless, the 30-day survival after OHCA resembles the previously described values of 9% [17,18]. This was achieved despite the fact that the present cohort may consist of a selected high-risk group in need of prolonged CPR as compared with patients who are easily resuscitated within a few minutes after start of CPR or defibrillation. The two patients admitted to hospital with on-going LUCAS CPR and who were treated with cardiopulmonary support were resuscitated during a particularly long time; regardless of this, both patients were alive after 30 days with minimal neurological sequelae. We believe that it is unlikely that these patients would have survived if transported and treated with manual CPR only. Furthermore, manual CPR is cumbersome during catheterisation and potentially hazardous for the health care personnel due to radiation exposure. Therefore, the LUCAS device seems ideal during catheterisation compared with manual CPR .
There are some limitations to this study. First, it is a small observational, prospective study with potentially unknown confounders; however, the relative advantage of this study design is that being their own controls, the patients studied are the same during manual CPR and LUCAS CPR. Second, the physician of the prehospital critical care team decides if the LUCAS device is to be used in the resuscitation, which undoubtedly introduces selection bias. Third, even though TTI and ECG data transfer to the central server was mandatory, a large part of the OHCA data was never transferred to the server.
Ventilation and compression depth measurements during resuscitation are recommended for the evaluation of CPR quality; however, these measurements cannot be assessed from TTI data alone, and little data support the clinical importance of these variables in this setting . The prehospital critical care teams were to take part in filling out a separate study form regarding the end-tidal CO2 (ETCO2), which is considered a surrogate measure of cardiac output during CPR ; however, little evidence exists to support this concept. Unfortunately, we are not able to report the ETCO2 values in this study due to the lack of registered values/too many missing values. Data on adverse device events or injuries have likewise not been reported in those patients not surviving OHCA due to the fact that these events were not described in the medical reports. The 45 patients admitted to hospital alive did not have any device-related events or injuries.
Mechanical chest compressions provided by the LUCAS device improve CPR quality by significantly reducing the NFF and by improving the quality of chest compression compared with manual CPR during OHCA resuscitation.
Randomised clinical studies not have been able to demonstrate improved survival for patients resuscitated with mechanical chest compression devices, and the devices have to some extent been discredited following the ASPIRE trial, the LINC trial and the PARAMEDIC trial. However, these trials are impeded by methodological deficiencies such as different resuscitation protocols in the compared groups and, in particular, prolonged time to deployment of the device. Maintenance of high-quality CPR during OHCA is not easy because of the small number of crew present, fatigue, patient access, and the impossibility of performing resuscitation in a moving vehicle. These considerations must be balanced against the current evidence from previous studies when deciding on the future role of mechanical CPR in healthcare systems, but it seems reasonably that such devices will continue to play a role when manual CPR is impractical. Nevertheless, it is essential that resources are available to support regular training of healthcare personnel ensuring that resuscitation guidelines are followed and to monitor CPR quality continuously.
The authors would like to thank all paramedics, nurses and physicians of the EMS, including the PMA and the HEMS for their collaboration in transferring TTI data. Physio-Control Inc. is thanked for providing the software necessary to analyse data and for providing equipment for the study.
The Danish Heart Foundation, The Riisfort Foundation, The Simon Fougner Hartmann Foundation, The Arvid Nielson Foundation and The Savvaerksejer Foundation supported the study.
- Edelson DP, Abella BS, Kramer-Johansen J, Wik L, Myklebust H, Barry AM, et al. Effects of compression depth and pre-shock pauses predict defibrillation failure during cardiac arrest. Resuscitation. 2006;71:137–45.View ArticlePubMedGoogle Scholar
- Deakin CD, Nolan JP, Soar J, Sunde K, Koster RW, Smith GB, et al. European Resuscitation Council Guidelines for Resuscitation 2010 Section 4. Adult advanced life support. Resuscitation. 2010;81:1305–52.View ArticlePubMedGoogle Scholar
- Wik L, Kramer-Johansen J, Myklebust H, Sorebo H, Svensson L, Fellows B, et al. Quality of cardiopulmonary resuscitation during out-of-hospital cardiac arrest. JAMA. 2005;293:299–304.View ArticlePubMedGoogle Scholar
- Krarup NH, Terkelsen CJ, Johnsen SP, Clemmensen P, Olivecrona GK, Hansen TM, et al. Quality of cardiopulmonary resuscitation in out-of-hospital cardiac arrest is hampered by interruptions in chest compressions-A nationwide prospective feasibility study. Resuscitation. 2011;82:263–9.View ArticlePubMedGoogle Scholar
- Bonnemeier H, Simonis G, Olivecrona G, Weidtmann B, Gotberg M, Weitz G, et al. Continuous mechanical chest compression during in-hospital cardiopulmonary resuscitation of patients with pulseless electrical activity. Resuscitation. 2011;82:155–9.View ArticlePubMedGoogle Scholar
- Wagner H, Madsen Hardig B, Steen S, Sjoberg T, Harnek J, Olivecrona GK. Evaluation of coronary blood flow velocity during cardiac arrest with circulation maintained through mechanical chest compressions in a porcine model. BMC Cardiovasc Disord. 2011;11:73.View ArticlePubMed CentralPubMedGoogle Scholar
- Rubertsson S, Lindgren E, Smekal D, Ostlund O, Silfverstolpe J, Lichtveld RA, et al. Mechanical chest compressions and simultaneous defibrillation vs conventional cardiopulmonary resuscitation in out-of-hospital cardiac arrest: the LINC randomized trial. JAMA. 2014;311:53–61.View ArticlePubMedGoogle Scholar
- Wik L, Olsen J, Persse D, Sterz F, Lozano Jr M, Brouwer MA, et al. Manual vs. integrated automatic load-distributing band CPR with equal survival after out of hospital cardiac arrest. The randomized CIRC trial. Resuscitation. 2014;85:741–8.View ArticlePubMedGoogle Scholar
- Axelsson C, Nestin J, Svensson L, Axelsson AB, Herlitz J. Clinical consequences of the introduction of mechanical chest compression in the EMS system for treatment of out-of-hospital cardiac arrest-a pilot study. Resuscitation. 2006;71:47–55.View ArticlePubMedGoogle Scholar
- Jacobs I, Nadkarni V, Bahr J, Berg RA, Billi JE, Bossaert L, et al. Cardiac arrest and cardiopulmonary resuscitation outcome reports: update and simplification of the Utstein templates for resuscitation registries. A statement for healthcare professionals from a task force of the international liaison committee on resuscitation (American Heart Association, European Resuscitation Council, Australian Resuscitation Council, New Zealand Resuscitation Council, Heart and Stroke Foundation of Canada, InterAmerican Heart Foundation, Resuscitation Council of Southern Africa). Resuscitation. 2004;63:233–49.View ArticlePubMedGoogle Scholar
- Stecher FS, Olsen JA, Stickney RE, Wik L. Transthoracic impedance used to evaluate performance of cardiopulmonary resuscitation during out of hospital cardiac arrest. Resuscitation. 2008;79:432–7.View ArticlePubMedGoogle Scholar
- Yost D, Phillips RH, Gonzales L, Lick CJ, Satterlee P, Levy M, et al. Assessment of CPR interruptions from transthoracic impedance during use of the LUCAS™ mechanical chest compression system. Resuscitation. 2012;83:961–5.View ArticlePubMedGoogle Scholar
- Kramer-Johansen J, Edelson DP, Losert H, Kohler K, Abella BS. Uniform reporting of measured quality of cardiopulmonary resuscitation (CPR). Resuscitation. 2007;74:406–17.View ArticlePubMedGoogle Scholar
- Hallstrom A, Rea TD, Sayre MR, Christenson J, Anton AR, Mosesso Jr VN, et al. Manual chest compression vs use of an automated chest compression device during resuscitation following out-of-hospital cardiac arrest: a randomized trial. JAMA. 2006;295:2620–8.View ArticlePubMedGoogle Scholar
- Putzer G, Braun P, Zimmermann A, Pedross F, Strapazzon G, Brugger H, et al. LUCAS compared to manual cardiopulmonary resuscitation is more effective during helicopter rescue-a prospective, randomized, cross-over manikin study. Am J Emerg Med. 2013;31:384–9.View ArticlePubMedGoogle Scholar
- Olasveengen TM, Wik L, Steen PA. Quality of cardiopulmonary resuscitation before and during transport in out-of-hospital cardiac arrest. Resuscitation. 2008;76:185–90.View ArticlePubMedGoogle Scholar
- Wissenberg M, Lippert FK, Folke F, Weeke P, Hansen CM, Christensen EF, et al. Association of national initiatives to improve cardiac arrest management with rates of bystander intervention and patient survival after out-of-hospital cardiac arrest. JAMA. 2013;310:1377–84.View ArticlePubMedGoogle Scholar
- Nichol G, Thomas E, Callaway CW, Hedges J, Powell JL, Aufderheide TP, et al. Regional variation in out-of-hospital cardiac arrest incidence and outcome. JAMA. 2008;300:1423–31.View ArticlePubMed CentralPubMedGoogle Scholar
- Wagner H, Terkelsen C, Friberg H, Harnek J, Kern K, Lassen JF, et al. Cardiac arrest in the catheterisation laboratory: a 5-year experience of using mechanical chest compressions to facilitate PCI during prolonged resuscitation efforts. Resuscitation. 2010;81:383–7.View ArticlePubMedGoogle Scholar
- Sheak KR, Wiebe DJ, Leary M, Babaeizadeh S, Yuen TC, Zive D, et al. Quantitative relationship between end-tidal carbon dioxide and CPR quality during both in-hospital and out-of-hospital cardiac arrest. Resuscitation. 2015;89C:149–54.View ArticleGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.