Monitoring of cerebral oxygen saturation during resuscitation in out-of-hospital cardiac arrest: a feasibility study in a physician staffed emergency medical system
© Schewe et al.; licensee BioMed Central Ltd. 2014
Received: 18 January 2014
Accepted: 29 September 2014
Published: 5 October 2014
Despite recent advances in resuscitation algorithms, neurological injury after cardiac arrest due to cerebral ischemia and reperfusion is one of the reasons for poor neurological outcome. There is currently no adequate means of measuring cerebral perfusion during cardiac arrest. It was the aim of this study to investigate the feasibility of measuring near infrared spectroscopy (NIRS) as a potential surrogate parameter for cerebral perfusion in patients with out-of-hospital resuscitations in a physician-staffed emergency medical service.
An emergency physician responding to out-of-hospital emergencies was equipped with a NONIN cerebral oximetry device. Cerebral oximetry values (rSO2) were continuously recorded during resuscitation and transport. Feasibility was defined as >80% of total achieved recording time in relation to intended recording time.
10 patients were prospectively enrolled. In 89.8% of total recording time, rSO2 values could be recorded (213 minutes and 20 seconds), thus meeting feasibility criteria. 3 patients experienced return of spontaneous circulation (ROSC). rSO2 during manual cardiopulmonary resuscitation (CPR) was lower in patients who did not experience ROSC compared to the 3 patients with ROSC (31.6%, ± 7.4 versus 37.2% ± 17.0). ROSC was associated with an increase in rSO2. Decrease of rSO2 indicated occurrence of re-arrest in 2 patients. In 2 patients a mechanical chest compression device was used. rSO2 values during mechanical compression were increased by 12.7% and 19.1% compared to manual compression.
NIRS monitoring is feasible during resuscitation of patients with out-of-hospital cardiac arrest and can be a useful tool during resuscitation, leading to an earlier detection of ROSC and re-arrest. Higher initial rSO2 values during CPR seem to be associated with the occurrence of ROSC. The use of mechanical chest compression devices might result in higher rSO2. These findings need to be confirmed by larger studies.
Spontaneous circulation in out of hospital cardiac arrest (OHCA) may be restored in up to 50% of patients in the presence of well-trained emergency physicians -. Despite these promising results in the treatment of OHCA, survival rates remain low.
Discharge rates of 14% and 20% are reported in these emergency medical systems (EMS) , and 1-year survival rates can reach up to 11% . Outcome depends on professional EMS treatment including proper post resuscitation therapy and implementation of treatment strategies as published in current guidelines . Lately, investigations revealed an improvement in neurological outcome in patients treated with therapeutic hypothermia following return of spontaneous circulation (ROSC) after OHCA . Multiple mechanisms are discussed to be responsible for this neuroprotective effect . Undoubtfully, sufficient perfusion pressure during CPR is also crucial for neurological outcome. During basic life support chest compressions (frequency and depth of compression) as well as ventilation (end-expiratory CO2) are poor measures to evaluate the performance of adequate resuscitation. During CPR, cerebral blood flow decreases to 20-50% compared to normal values -. At the same time, neurological outcome and survival depend on sufficient cerebral blood flow. To date, it is impossible to provide helpful measurements to predict neurological outcome during out-of-hospital resuscitation.
Regional cerebral oximetry with near infrared spectroscopy (NIRS) has emerged as a surrogate parameter monitoring cerebral perfusion in the intraoperative and intensive care setting . The method uses the effect that light in the near-infrared spectrum can penetrate the skull, thus allowing measurements of oxyhemoglobin. The absorption of light permits the measurement of oxyhemoglobin, desoxyhemoglobin, and total hemoglobin . Ono et al. have revealed that decreasing intraoperative rSO2 values due to hypotension are associated with major morbidity and mortality after cardiac surgery . Murkin et al. could show that intraoperative treatment of low rSO2 values resulted in decreased major organ morbidity or mortality . Small clinical studies have addressed the feasibility of NIRS technology in measuring cerebral oxygen saturation in patients during CPR. These studies included in-hospital resuscitation -, and out-of-hospital cardiac arrest ,, as well as a combination of the two ,. The authors suggest that the use of NIRS for cerebral oximetry is promising for monitoring patients with cardiac arrest. However, none of these studies focused on routine use in a physician-staffed EMS. In summary, data is still limited, but available studies suggest that NIRS monitoring can provide a real-time non-invasive marker of cerebral perfusion and thus cerebral oximetry may have a role in optimising cerebral perfusion in cardiac arrest.
The aim of this study was to investigate the feasibility of NIRS monitoring in out-of-hospital cardiac arrests in a physician-staffed emergency medical service in daily routine. Cerebral oximetry (rSO2) was measured during standard cardiopulmonary resuscitation. This included the optional use of a mechanical chest compression device, a decision left to the discretion of the emergency physician.
The emergency medical service (EMS) of the City of Bonn serves 320,000 residents in a service area of 141 km2, with a population density of 2,250 persons/km2. The city reflects urban features. A total of 12 basic life support (BLS) units and 3 advanced life support (ALS) units serve (2 of them 24 hours/7 days and 1 only 10 hours weekdays) in a two tier system. The EMS has 84,240 BLS-unit hours/year and 19.410 ALS-unit hours/year.
The EMS system responds to 33,600 emergency calls, increasing about 3-5% per year, with 26,000 BLS-unit transports and 7,600 ALS unit interventions per year. In Bonn patients are admitted to 12 hospitals, all with different care levels.
The BLS units are at least staffed with one emergency medical technician (EMT) and one paramedic. The ALS units are staffed with one paramedic and one physician. Physicians on the ALS units are predominantly anaesthesiologists and have passed at least 2 years of postgraduate training as well as a special emergency training course according to regulations of the German medical academy. The incidence of CPR in the EMS of Bonn is within 45–55 CPRs/100,000 inhabitants/year.
The NONIN EQUANOX Model 7600 regional oximetry system (Nonin Medical, Plymouth, Minnesota, USA) is a portable 4-wavelength cerebral oximeter. It weighs approximately 900 grams, plus 180 grams for 2 sensor pods in the 2 sensor configuration. Dimensions are 305×108×130 mm. Once the device is activated and a sensor is attached, it displays continuous rSO2 data in percent from 0 to 100. Readings are sampled every 4 seconds. The monitor is equipped with a storage battery lasting for 4 hours when fully charged.
Study population/data acquisition
The study was performed from October 2012 to July 2013. One ALS unit, which is on duty from Monday until Friday from 8 a.m.- 6 p.m., was equipped with the NIRS device. Prior to the beginning of the study, 3 physicians were trained for half an hour in the use of the device according to legal requirements. Whenever dispatched to emergencies with suspected cardiac arrest, which was the case when a lifeless person was reported, the physician carried the NIRS device to the scene. All patients age >18 years with non-traumatic cardiac arrest were included. Exclusion criteria were: trauma, hypothermia <35°C, and pregnancy.
If the inclusion criteria were met, the physician immediately initiated the NIRS monitoring at the time of beginning of ALS. One optode was attached on the patient's left forehead lateral of the midline and above the eyebrow to acquire the spectrum of the left frontal cortex. This process took less than 30 seconds and did not interrupt BLS. Time markers were set on the device for important events such as ROSC, use of mechanical compression devices, or termination of CPR. Advanced cardiac life support (ACLS) was performed according to 2010 ERC guidelines . This included intubation by the emergency physician during ALS verified by direct laryngoscopy, auscultation and end-tidal CO2 measurement. Ventilator settings were adjusted according to the patient's estimated body weight. The rSO2 value on the device monitor was covered for the study, so that the resuscitation team was blinded to rSO2 readings and the process of routine resuscitation was not altered by the NIRS device. Intended recording time of rSO2 started immediately after the emergency physician arrived at the scene with initiation of ALS and ended when the patient arrived at the emergency department or when CPR was terminated and death testfied. Data was transferred onto a computer via bluetooth and data was analyzed offline on a standard Excel spreadsheet also used for descriptive statistics. We defined feasibility as percentage (>80%) of total achieved recording time in relation to intended recording time. Data is displayed as mean ± standard deviation if not mentioned otherwise.
The study was approved by the local ethics committee of the University of Bonn Medical Center.
Demographic and NIRS data of all 10 included patients
rSO2during manual CPR (%)
rSO2during mechanical compression (%)
rSO2after ROSC (%)
rSO2after end of CPR (no ROSC, %)
35.3 ± 6.9
58.3 ± 4.6
35.4 ± 3.0
33.1 ± 0.9
30.5 ± 0.7
30.9 ± 0.3
35.6 ± 1.7
36.0 ± 0.4
45.7 ± 5.3
63.2 ± 8.0
38.0 ± 5.9
29.3 ± 1.4
18.2 ± 3.1
20.4 ± 0.5
30.5 ± 1.3
35.3 ± 4.2
37.4 ± 4.7
42.2 ± 2.9
21.3 ± 2.5
25.4 ± 2.3
In all patients with ROSC (n = 3), an increase in rSO2 was observed under spontaneous circulation. Mean rSO2 increased from 37.2% ± 7.7 before ROSC to 52.3% ± 14.9 at time of ROSC, a relative increase of 40.6%.
Offline analysis showed that ROSC was indicated by an increase in rSO2 well before the emergency physician diagnosed ROSC through a palpable pulse in 2 patients (Figure 2). In 1 patient with initial asystole rSO2 increased after ROSC was diagnosed (Figure 3). In parallel, rSO2 decreased before the emergency physisican decided to restart CPR due to recurrent ventricular fibrillation in 2 patients (see raw data of patient 1, Figure 2).
This study demonstrated the feasibility of rSO2 monitoring via NIRS during resuscitation in out-of-hospital cardiac arrest in a physician-staffed emergency medical system, both at the scene and also during transport of patients. Feasibility criteria for monitoring rSO2 were met by the NIRS device, providing values in 89.8% of total recording time. Moreover, a concomitant detection of increasing NIRS values in patients where ROSC was obtained and a decrease of NIRS values following re-arrest with the onset of VT was observed.
In a different study, Newman et al. reported no detectable rSO2 signal in patients with out-of-hospital cardiac arrest . Contrary to these findings, we could record stable values, possibly due to a general technological improvement in modern NIRS devices.
Only 1 out of 10 patients survived without neurological impairment (Cerebral Performance Category, CPC 1, good performance). This conforms with the general resuscitation outcome of the EMS of the City of Bonn as reported in a larger study . It is remarkable that this patient had the highest rSO2 readings during CPR and ROSC (45.7% ± 5.27 and 63.2%, ± 8.02, respectively). However, this individual finding is too limited to draw conclusions on general neurological outcome and NIRS measurements during CPR.
Ahn et al. suggested a relationship between rSO2 and ROSC as well as a relationship between low rSO2 and unsuccessful CPR  Koyama et al. drew similar conclusions following their results . Our results are consistent with these previous findings, observing lower rSO2 readings corresponding with futile resuscitation efforts compared to rSO2 values in patients with successful CPR (rSO2 31.5% ± 7.1 vs. 37.2% ± 17.0, Figure 5). The second patient to experience ROSC survived more than 24 hours, but ultimately suffered hypoxic brain damage. In this patient, ROSC was detected after a notable increase in rSO2 (35.3%, ± 6.94 during CPR vs. 58.3%, ± 4.63 after ROSC). However, before ROSC occurred, NIRS readings showed a long period of low rSO2 (Figure 2). One could speculate rSO2 levels were low for too long and neurological damage was therefore irreversible. Unfortunately, NIRS monitoring was not continued after hospital admission. In future, the initial NIRS reading and the trend during the initial phase of resuscitation might help to predict outcome in terms of ROSC, survival and/or neurological outcome.
In the 2 patients mentioned above, rSO2 decreased before cardiac re-arrest was observed by the emergency physician. Frisch et al. published similar observations for peripheral tissue oximetry  and Meex et al. for cerebral oximetry . We believe that NIRS could potentially improve CPR protocols by serving as an early-warning system detecting hemodynamic instability after ROSC prior to re-arrest.
In both of these patients, NIRS values increased rapidly with ROSC. In contrast, rSO2 increased slowly after ROSC in patient 8 (see Figure 3). Vasopressors had to be used to sustain circulation during transportation until therapy was finally withdrawn in the emergency department. Our study is too limited to draw assumptions between the dynamics of increase in NIRS following ROSC and stable versus unstable hemodynamics and the possible underlying pathology.
Several studies have found an increase in rSO2 after ROSC in patients -. Only Kämärainen et al. reported that rSO2 increases prior to clinical detection of ROSC . It may be speculated that the time delay between increase of rSO2 in 2 cases of our study prior to documented ROSC is due to delayed clinical detection of ROSC by the physician. Guidelines recommend continuous chest compression for 2 minutes after defibrillation before analysis of cardiac rhythm and pulse. Nevertheless, within these two minutes, ROSC might be detected by increasing rSO2. In future, it may be possible to differentiate between sufficient and insufficient cerebral perfusion as detected by a significant increase in NIRS guiding the physician to continue or interrupt manual chest compression.
The use of mechanical compression devices for CPR still remains controversial. In previous studies, a higher arterial pressure and cardiac perfusion pressure were obtained with mechanical CPR devices versus manual chest compression ,. A recent meta-analysis by Westfall et al. suggests the superiority of mechanical devices versus manual chest compression regarding ROSC . However, there is insufficient evidence for the superiority of mechanical chest compression devices in terms of neurological outcome . To our knowledge, we are the first to provide data on NIRS monitoring during mechanical and manual chest compression in OHCA. Kämärainen et al. could not show a significant increase in rSO2 by improving CPR with a feedback device (improved quality of manual chest compression) . Whether this also applies to mechanical CPR devices has not been shown. In both patients where the device was applied, rSO2 values were increased by 12% and 19% compared to manual compression. This finding could support the theory that the use of mechanical CPR devices can achieve higher cerebral perfusion pressure compared to manual chest compression. However, we cannot rule out that increasing arterial CO2 might have influenced measured rSO2 values based on autoregulation of cerebral blood flow influenced by changing arterial CO2. The detected increase in measured rSO2 values during mechanical chest compressions could in theory also be explained by increasing arterial CO2.
During resuscitation, end-tidal CO2 values are also used as a surrogate parameter for arterial CO2 and thus as an indirect measurement of the achieved cardiac output during CPR. End-tidal CO2 during resuscitation is believed to be a quality marker of CPR performance . In this study, end-tidal CO2 values during resuscitation were measured, but not recorded to the memory card due to technical reasons, and thus were not available for offline analysis, which limits our observational study.
Cerebral oximetry using NIRS for out-of-hospital cardiac arrest is feasible in a physician-staffed EMS. rSO2 monitoring can help detect ROSC and hemodynamic instabilities resulting in re-arrests. rSO2 values measured during application of a mechanical chest compression device were higher than during manual chest compression. Further investigations are needed to confirm whether these results can be explained by improved cerebral perfusion.
We thank the emergency physicians and the Fire Department of the City of Bonn for their continued support.
- Berdowski J, Berg RA, Tijssen JGP, Koster RW: Global incidences of out-of-hospital cardiac arrest and survival rates: Systematic review of 67 prospective studies. Resuscitation. 2010, 81: 1479-1487. 10.1016/j.resuscitation.2010.08.006.View ArticlePubMedGoogle Scholar
- Fischer M, Fischer NJ, Schüttler J: One-year survival after out-of-hospital cardiac arrest in Bonn city: outcome report according to the "Utstein style". Resuscitation. 1997, 33: 233-243. 10.1016/S0300-9572(96)01022-2.View ArticlePubMedGoogle Scholar
- Herlitz J, Bahr J, Fischer M, Kuisma M, Lexow K, Thorgeirsson G: Resuscitation in Europe: a tale of five European regions. Resuscitation. 1999, 41: 121-131. 10.1016/S0300-9572(99)00045-3.View ArticlePubMedGoogle Scholar
- Neukamm J, Gräsner J-T, Schewe J-C, Breil M, Bahr J, Heister U, Wnent J, Bohn A, Heller G, Strickmann B, Fischer H, Kill C, Messelken M, Bein B, Lukas R, Meybohm P, Scholz J, Fischer M: The impact of response time reliability on CPR incidence and resuscitation success: a benchmark study from the German Resuscitation Registry. Crit Care Lond Engl. 2011, 15: R282-10.1186/cc10566.View ArticleGoogle Scholar
- Nolan JP, Soar J, Zideman DA, Biarent D, Bossaert LL, Deakin C, Koster RW, Wyllie J, Böttiger B: European Resuscitation Council Guidelines for Resuscitation 2010 Section 1: Executive summary. Resuscitation. 2010, 2010 (81): 1219-1276. 10.1016/j.resuscitation.2010.08.021.View ArticleGoogle Scholar
- Arrich J, Holzer M, Havel C, Müllner M, Herkner H: Hypothermia for neuroprotection in adults after cardiopulmonary resuscitation. Cochrane Database Syst Rev. 2012, 9: CD004128-PubMedGoogle Scholar
- Chalkias A, Xanthos T: Post-cardiac arrest brain injury: pathophysiology and treatment. J Neurol Sci. 2012, 315: 1-8. 10.1016/j.jns.2011.12.007.View ArticlePubMedGoogle Scholar
- Pernat A, Weil MH, Sun S, Tang W: Stroke volumes and end-tidal carbon dioxide generated by precordial compression during ventricular fibrillation. Crit Care Med. 2003, 31: 1819-1823. 10.1097/01.CCM.0000069538.12447.82.View ArticlePubMedGoogle Scholar
- Bartlett RL, Stewart NJ, Raymond J, Anstadt GL, Martin SD: Comparative study of three methods of resuscitation: closed-chest, open-chest manual, and direct mechanical ventricular assistance. Ann Emerg Med. 1984, 13 (9 Pt 2): 773-777. 10.1016/S0196-0644(84)80433-3.View ArticlePubMedGoogle Scholar
- Fischer M, Dahmen A, Standop J, Hagendorff A, Hoeft A, Krep H: Effects of hypertonic saline on myocardial blood flow in a porcine model of prolonged cardiac arrest. Resuscitation. 2002, 54: 269-280. 10.1016/S0300-9572(02)00151-X.View ArticlePubMedGoogle Scholar
- Ghosh A, Elwell C, Smith M: Review article: cerebral near-infrared spectroscopy in adults: a work in progress. Anesth Analg. 2012, 115: 1373-1383. 10.1213/ANE.0b013e31826dd6a6.View ArticlePubMedGoogle Scholar
- Moerman A, Wouters P: Near-infrared spectroscopy (NIRS) monitoring in contemporary anesthesia and critical care. Acta Anaesthesiol Belg. 2010, 61: 185-194.PubMedGoogle Scholar
- Ono M, Brady K, Easley RB, Brown C, Kraut M, Gottesman RF, Hogue CW: Duration and magnitude of blood pressure below cerebral autoregulation threshold during cardiopulmonary bypass is associated with major morbidity and operative mortality. J Thorac Cardiovasc Surg. 2014, 147: 483-489. 10.1016/j.jtcvs.2013.07.069.View ArticlePubMedGoogle Scholar
- Murkin JM, Adams SJ, Novick RJ, Quantz M, Bainbridge D, Iglesias I, Cleland A, Schaefer B, Irwin B, Fox S: Monitoring brain oxygen saturation during coronary bypass surgery: a randomized, prospective study. Anesth Analg. 2007, 104: 51-58. 10.1213/01.ane.0000246814.29362.f4.View ArticlePubMedGoogle Scholar
- Kämäräinen A, Sainio M, Olkkola KT, Huhtala H, Tenhunen J, Hoppu S: Quality controlled manual chest compressions and cerebral oxygenation during in-hospital cardiac arrest. Resuscitation. 2012, 83: 138-142. 10.1016/j.resuscitation.2011.09.011.View ArticlePubMedGoogle Scholar
- Ito N, Nanto S, Nagao K, Hatanaka T, Nishiyama K, Kai T: Regional cerebral oxygen saturation on hospital arrival is a potential novel predictor of neurological outcomes at hospital discharge in patients with out-of-hospital cardiac arrest. Resuscitation. 2012, 83: 46-50. 10.1016/j.resuscitation.2011.10.016.View ArticlePubMedGoogle Scholar
- Koyama Y, Wada T, Lohman BD, Takamatsu Y, Matsumoto J, Fujitani S, Taira Y: A new method to detect cerebral blood flow waveform in synchrony with chest compression by near-infrared spectroscopy during CPR. Am J Emerg Med. 2013, 31: 1504-1508. 10.1016/j.ajem.2013.07.002.View ArticlePubMedGoogle Scholar
- Newman DH, Callaway CW, Greenwald IB, Freed J: Cerebral oximetry in out-of-hospital cardiac arrest: standard CPR rarely provides detectable hemoglobin-oxygen saturation to the frontal cortex. Resuscitation. 2004, 63: 189-194. 10.1016/j.resuscitation.2004.05.003.View ArticlePubMedGoogle Scholar
- Frisch A, Suffoletto BP, Frank R, Martin-Gill C, Menegazzi JJ: Potential utility of near-infrared spectroscopy in out-of-hospital cardiac arrest: an illustrative case series. Prehospital Emerg Care Off J Natl Assoc EMS Physicians Natl Assoc State EMS Dir. 2012, 16: 564-570. 10.3109/10903127.2012.702191.View ArticleGoogle Scholar
- Meex I, Deyne CD, Dens J, Scheyltjens S, Lathouwers K, Boer W, Vundelinckx G, Heylen R, Jans F: Feasibility of absolute cerebral tissue oxygen saturation during cardiopulmonary resuscitation. Crit Care. 2013, 17: R36-10.1186/cc12546.PubMed CentralView ArticlePubMedGoogle Scholar
- Ahn A, Nasir A, Malik H, D’Orazi F, Parnia S: A pilot study examining the role of regional cerebral oxygen saturation monitoring as a marker of return of spontaneous circulation in shockable (VF/VT) and non-shockable (PEA/Asystole) causes of cardiac arrest. Resuscitation. 2013, 84: 1713-1716. 10.1016/j.resuscitation.2013.07.026.View ArticlePubMedGoogle Scholar
- Duchateau F-X, Gueye P, Curac S, Tubach F, Broche C, Plaisance P, Payen D, Mantz J, Ricard-Hibon A: Effect of the AutoPulse automated band chest compression device on hemodynamics in out-of-hospital cardiac arrest resuscitation. Intensive Care Med. 2010, 36: 1256-1260. 10.1007/s00134-010-1784-x.PubMed CentralView ArticlePubMedGoogle Scholar
- Timerman S, Cardoso LF, Ramires JAF, Halperin H: Improved hemodynamic performance with a novel chest compression device during treatment of in-hospital cardiac arrest. Resuscitation. 2004, 61: 273-280. 10.1016/j.resuscitation.2004.01.025.View ArticlePubMedGoogle Scholar
- Westfall M, Krantz S, Mullin C, Kaufman C: Mechanical versus manual chest compressions in Out-of-hospital cardiac arrest: a meta-analysis. Crit Care Med. 2013, 41: 1782-1798. 10.1097/CCM.0b013e31828a24e3.View ArticlePubMedGoogle Scholar
- Ong MEH, Mackey KE, Zhang ZC, Tanaka H, Ma MH-M, Swor R, Shin SD: Mechanical CPR devices compared to manual CPR during out-of-hospital cardiac arrest and ambulance transport: a systematic review. Scand J Trauma Resusc Emerg Med. 2012, 20: 39-10.1186/1757-7241-20-39.PubMed CentralView ArticlePubMedGoogle Scholar
- Kolb JC, Ainslie PN, Ide K, Poulin MJ: Effects of five consecutive nocturnal hypoxic exposures on the cerebrovascular responses to acute hypoxia and hypercapnia in humans. J Appl Physiol Bethesda Md 1985. 2004, 96: 1745-1754.Google Scholar
- Morley PT: Monitoring the quality of cardiopulmonary resuscitation. Curr Opin Crit Care. 2007, 13: 261-267. 10.1097/MCC.0b013e32814b05bd.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.