- Open Access
The measurement of tissue interface pressures and changes in jugular venous parameters associated with cervical immobilisation devices: a systematic review
Scandinavian Journal of Trauma, Resuscitation and Emergency Medicine volume 21, Article number: 81 (2013)
Cervical immobilisation is commonly applied following trauma, particularly blunt head injury, but current methods of immobilisation are associated with significant complications. Semi-rigid disposable cervical collars are known to cause pressure ulcers, and impede effective airway management. These collars may also exacerbate a head injury by increasing intracranial pressure as a result of external compression of the jugular veins. There is a clear imperative to find ways of effectively immobilising the cervical spine whilst minimising complications, and any assessment of existing or new devices should include a standardized approach to the measurement of tissue interface pressures and their effect on jugular venous drainage from the brain. This systematic review summarises the research methods and technologies that have been used to measure tissue interface pressure and assess the jugular vein in the context of cervical immobilisation devices. 27 papers were included and assessed for quality. Laboratory investigations and biomechanical studies have gradually given way to methods that more accurately reflect clinical care. There are numerous accounts of skin ulceration associated with cervical collars, but no standardised approach to measuring tissue interface pressure. It is therefore difficult to compare studies and devices, but a pressure of less than 30 mmHg appears desirable. Cervical collars have been shown to have a compressive effect on the jugular veins, but it is not yet certain that this is the cause of the increased intracranial pressure observed in association with cervical collar use. This is the first review of its type. It will help guide further research in this area of trauma care, and the development and testing of new cervical immobilisation devices.
Traumatic injuries as a result of road traffic collisions are becoming more prevalent worldwide, with over a million deaths annually, and this figure is predicted to rise . In developed countries it is standard practice both in pre-hospital and in-hospital care to apply a semi-rigid cervical collar to the neck of patients presenting with blunt trauma, particularly head injury. These collars are applied with the aim of minimising neck movement prior to definitive clinical and/or radiological assessment, in order to reduce the risk of further damage that may result if unrestricted neck movement occurs in the presence of an unstable cervical spine injury. This practice is currently adopted by all ambulance trusts in the United Kingdom (UK), and promoted by the Advanced Trauma Life Support (ATLS) and similar courses .
However, the actual incidence of spinal cord injury is very low, at 10-15 per million population per year in the UK . In the vast majority of patients semi-rigid cervical collars are fitted “as a precaution” in the absence of cervical spine injury; this means that most patients do not benefit from their application. Semi-rigid collars are often left in place for many hours, particularly in unconscious patients, with on-going use on intensive care and orthopaedic or neurosurgical wards. In a healthy and non-spinally injured patient continuous contact with the collar, and the resulting tissue interface pressures, can lead to early tissue breakdown and ulceration . This is even more problematic in the presence of an actual spinal injury, due to lack of neural tone and decreased blood flow to the affected area . As a result, the onset of ulceration and tissue breakdown is more rapid in patients where it is least desirable.
It seems likely that that the use of semi-rigid collars can exacerbate an existing traumatic brain injury [6–8]. This potential to exacerbate a head injury appears to occur as a result of increases in intracranial pressure (ICP) associated with the application of a semi-rigid collar, but is not well understood. Existing research suggests that increases in ICP are attributable to some degree of interaction with the jugular veins [9, 10]. However, further work is required in this area.
Since most patients to whom a semi-rigid collar is applied do not need cervical immobilisation, and with such devices known to cause clinically significant complications, new procedures and devices are clearly required . Research methods regarding cervical range of motion have recently been reviewed ; this should be taken into consideration when conducting a thorough evaluation of devices. In addition, any assessment of existing or new devices should include a standardized approach to the measurement of tissue interface pressure and the devices effect on jugular venous drainage from the brain. We therefore undertook a systematic review to summarise the evidence to date and inform further research in this area.
This systematic review was conducted to identify and summarise research relevant to the measurement of tissue interface pressures and assessment of the jugular veins and intracranial pressure, with reference to cervical immobilisation devices. The review has two objectives:
To identify scientific methods used to measure tissue interface pressures occurring in association with cervical immobilisation devices and summarise the findings from the research;
To identify scientific methods used to research the jugular vein and intracranial pressure in association with cervical immobilisation devices and summarise findings from the research.
Data sources and search strategy
A systematic search of international literature was conducted in AMED (Allied and Complementary Medicine), BNI (British Nursing Index), EMBASE (biomedical and pharmacological), MEDLINE (biomedical and life sciences), and CINAHL (nursing, biomedicine and health sciences) databases. A qualified librarian assisted in planning and performing the searches. Boolean/phase mode was applied for two searches and the search strategy is shown in Appendix 1. It was necessary to conduct a key word search as there is no MeSH term for cervical collar or orthoses. The searches were restricted to papers published in the English language and were conducted during October 2012; the last access to the database was on 31st October 2012.
Inclusion and exclusion criteria
Scoping searches revealed that there were likely to be only a limited number of relevant randomised studies. Therefore, quasi-experimental and observational studies were eligible for inclusion. Due to the paucity of research in this area case reports and literature reviews were also included.
Articles were excluded if the primary focus of the paper was children, animals, chronic neck complaints or pressure ulcers caused by devices not designed to immobilise the cervical spine.
The study selection was a two stage process. Study titles and, where available, abstracts were initially screened by a single reviewer (AS) and first stage decision against the exclusion criteria was made. All relevant full-text articles were retrieved for detailed assessment by two reviewers (AS and SV) and those meeting the exclusion criteria were rejected. Studies relevant to the objectives were included in the final review.
Each paper was assessed for quality and the limitations of each study, including risk of bias, were considered. These are summarised in Tables 1 and 2. This area of research is currently in development and there is a lack of published research. None of the papers returned by the searches were large prospective randomised controlled trials assessing interface pressures or changes in jugular venous parameters associated with cervical immobilisation devices. Therefore, no formal risk of bias tools were used. However, it is important to fully consider the impact of any limitations to inform future investigations. For this reason all studies identified were included in the review and the various limitations of the studies are discussed in the results.
116 articles were returned from the search. 81 were excluded in the first stage of assessment and a further 7 during the second stage. 27 papers were included in the review.
Information was extracted from each study on tissue interface pressures (14 papers included [4, 5, 10, 13–23]) and jugular venous measurements (14 papers included [6–10, 24–32]) in association with cervical immobilisation devices (Figure 1). One paper  was included in both searches.
Research and scientific methods to measure tissue interface pressures
There have been a number of observational, retrospective studies to document cases of skin breakdown and ulceration resulting from semi-rigid cervical collars. The 14 papers included in this section of the review are summarised in Table 1.
In 1994 Liew et al carried out a case review of two multi-trauma patients , noting that they had developed significant occipital pressure ulceration as a result of cervical collar use, whilst Murphy, in 1997 , was the first to note that ‘development of occipital pressure ulcers under cervical collars is a common problem’ (Pg. 60). Molano et al also identified pressure sores as a complication of collar use following a review of patients admitted with an acute cervical spine injury . Pressure sores developed most frequently at occipital, chin and supra-scapular zones; with occipital injuries the most severe.
Jacobsen et al determined that whilst only 1% of all occipital pressure ulcers are attributable to the use of a rigid cervical collar , the incidence of hospital acquired pressure ulcer ranges from 23.9% to 44%. They also noted that collar-related pressure ulcers may develop at various locations on the body; but occipital pressure ulcers are the most serious because there is very little subcutaneous tissue over the bone, confirming the work of Molano et al . Jacobsen et al also drew attention to cases where the application of cervical collars exerted a pressure greater than the capillary closing pressure, thus restricting blood flow to contact areas, exacerbating the problem. A retrospective case review by Walker identified immobilised patients as being at greatest risk of pressure ulceration exacerbated by immobilisation devices . This resulted in prolonged recovery times and additional healthcare dependence. He recommended that practitioners be aware of the high risk of pressure ulcer development in patients with a cervical orthosis, especially if they are not able to relieve the pressure or verbalise discomfort. Walker also considered the factors that affect development of pressure ulcers in the spinally injured immobilised patient, such as reduced mobility, use of immobilisation devices, malnutrition and hypoperfusion.
Following the observation that skin breakdown and ulceration are common after the application of a cervical collar, the next logical step became measurement of the tissue interface pressures that occur.
Fisher observed that the resting tissue interface pressure (TIP) exerted by a sternal occipital mandibular immobilisation (SOMI) orthosis depended on how tight the orthosis was fitted (pressures of between 25 and 105 mmHg were recorded) . However, when Beavis  and Ferguson et al  investigated TIP using pneumatic pressure sensors placed on the necks of volunteers, they obtained varied results. Ferguson et al recorded mean TIPs of between 0.3 and 11.8 mmHg from passive supine participants, whilst Beavis recorded TIPs of between 55 and 150 mmHg for active movements of flexion and extension with seated participants. The findings from these studies indicate that there is considerable variation in TIP measurements depending on the testing conditions and subject movement, with a lack of standardization in the testing protocols used.
Plaisier et al carried out a study comparing four cervical collars (Stifneck®, Miami J, Philadelphia and Newport) . Tissue interface pressures were measured using electro-pneumatic sensors at the chin, occiput and mandible, and opinions on comfort noted. All collars were tested on healthy volunteers in both a seated and supine position (adopting features from the work of Beavis and Ferguson et al). TIP exerted by the Stifneck® collar was noted to exceed capillary closing pressure (CCP); this also applied to the Philadelphia collar, particularly when wearers were supine. The Miami J and Newport exerted a TIP well below CCP. Newport and Miami J devices were therefore perceived as ‘skin friendly’, due to comfort and low TIP readings, with the conclusion that they may reduce the incidence of soft tissue complications and improve patient compliance.
Blaylock conducted a retrospective analysis of pressure ulcer development as well as a clinical trial of a new collar in 20 patients . The recommended outcomes were for improved skin care, education on proper collar fitting and choice of collar, as well as implementing the new collar for trauma patients. A multidisciplinary approach to reducing complications associated with collar use was also adopted by Powers . By improving nursing care, using a different type of collar and developing a new protocol for cervical spine clearance, Powers was able to eliminate the incidence of pressure ulcers related to collar use from his department. A key outcome from this work was that patients were wearing collars for a much shorter period of time. A recommendation for early collar removal was also the conclusion of a study by Chendrasekhar et al . They examined patients with severe closed head injury; 13 (34%) developed occipital pressure sores related to cervical collar use, whilst those who underwent early collar removal did not.
Black et al evaluated skin breakdown in healthy volunteers with two collars (Aspen and Philadelphia). They measured occipital pressure using the Talley digital skin pressure evaluator, Model SD500 (Talley Group Ltd, England); as well as skin temperature and humidity . Pressure and skin temperature were comparable between the collars, but skin humidity increased with the Philadelphia collar, increasing the risk of pressure ulcer development. Tescher et al carried out a similar study looking at the TIP exerted by four cervical collars (Miami J, Aspen, Philadelphia and Miami J) . Interface pressures were measured at the occiput and the anterior mandible using the XSensor X2 (XSENSOR Technology Corporation, Canada) system of capacitance pressure-sensing transducers, in both a seated and supine position, similar to Plaisier. They showed that the TIP exerted by the collar at the mandible varied greatly (between 22.08 and 70.89 mmHg); but more significantly they noted capillary closing pressure was only 32 mmHg. As the TIP sites were different it is difficult to compare these results to other studies, however Tescher does confirm Black’s theory that moisture may impact on pressure sore development. Tescher also identified a number of other important factors that influence TIP. These include BMI, fluid retention due to the traumatic inflammatory response, poorly fitting collars and shear forces exerted on the skin.
Research and scientific methods to assess the jugular vein and intracranial pressure
There has been a limited amount of work to assess the jugular vein and its relationship to intracranial pressure (ICP) when a cervical collar is applied. The link between the use of rigid cervical collars and an increase in intracranial pressure (especially in head injured patients) is widely considered as fact, yet the exact aetiology is unknown. The 14 papers included in this section of the review are summarised in Table 2.
A number of studies have postulated a venous tourniquet effect from a rigid cervical collar, with an impact on intracranial pressure [10, 24]. This has been supported by research demonstrating that ICP increases when a cervical collar is applied, and then gradually declines on removal. The most obvious explanation is an increase in intracranial blood volume due to pressure on the neck impairing venous drainage [6, 25–28].
Hunt et al  examined the effect of cervical collars on patients with a traumatic brain injury. Results showed a rise in baseline intracranial pressure when collars were applied. There were no other changes to cardio-respiratory parameters; therefore venous compression in the neck was considered to be the likely cause of the observed effect. Moreover, they showed that the most significant increases in intracranial pressure were in patients with a higher baseline to start with, suggesting that this effect of cervical collars is more pronounced in patients with a greater risk of neurological deterioration. These findings were confirmed by Mobbs et al who concluded that early assessment of the cervical spine in a head injured patient is essential . They also postulated that obstruction of venous drainage and the potential for the collar to cause a painful stimulus may both impact on ICP. However, they were unable to determine the exact reason for the observed increase in ICP.
Kuhnigk et al measured ICP in 18 severely head injured patients before and after placement of a collar , but did not find a significant change in ICP following collar application. They therefore concluded that the risk of increasing ICP in a head injured patient was low. These findings contradict the other papers reviewed, and are unexplained. However, the work was carried out in patients with a higher baseline ICP than the other studies, and this may have been a factor.
Stone et al were the first to actually assess the effect of a cervical collar on the dimensions of the internal jugular veins, using ultrasound examination . In healthy volunteers, they were able to show an increase in cross-sectional area following the application of a cervical collar, supporting the hypothesis that collars create venous obstruction in the neck. It seems likely that this obstruction contributes to an increase in intracranial pressure, but this could not be confirmed because the measurement of ICP is invasive, and therefore not suitable for volunteer studies.
A literature review by Ho et al noted that between 25% and 75% of all traumatic cervical spine injuries are unstable, and therefore recommended cervical stabilisation . However, they hypothesised that a cervical collar may act as a tourniquet around a patient’s neck, mirroring Ferguson’s concerns. They therefore recommended rapid clearance of the cervical spine in trauma patients to reduce the risk of raised intracranial pressure. Prolonged collar use was also the focus of a literature review by Dunham et al . The risk of raised ICP due to collar use was found to be 35.8% in unstable patients, and they also noted that a cervical collar was associated with increased intracranial pressure in patients without head injury. In a later simulation study, Dunham et al showed that early collar removal was associated with more favourable outcomes in both unstable and stable cervical spine injured patients .
This systematic review has identified relatively few studies that assess the impact of a semi-rigid cervical collar on tissue interface pressures or jugular venous drainage from the brain. Those that have been done tend to use small samples of volunteers with variable methodologies and testing protocols, which makes comparison between papers difficult.
The evidence that is available demonstrates an important causal relationship between the use of semi-rigid cervical collars and the development of pressure ulcers, as a result of the skin pressure exerted at key tissue interfaces. A number of retrospective studies have linked cervical collars to the development of pressure ulcers, but few have actually determined the TIP exerted by various immobilisation devices. For those that do record TIP the results vary widely, ranging from 0.3 mmHg to 150 mmHg, and this makes it difficult to draw any clear conclusions or compare different devices. Much of the variation is likely to be due to differences in test conditions (e.g. seated versus supine) and the technical characteristics of the devices used to measure TIP, as well as actual differences between the immobilisation devices under test. Future research in this area would benefit from the development and implementation of standardised TIP measurement techniques and protocols.
With respect to jugular vein assessment, we found a number of studies demonstrating an association between the use of semi-rigid cervical collars and increased intracranial pressure (ICP), though this was not universal. There is, however, very little research to determine the underlying cause. It seems likely that existing cervical collars have a tendency to obstruct jugular venous drainage, by acting as a tourniquet around the neck, but although it has been shown that collars have an effect on the dimensions of the internal jugular vein , a causal link to increased ICP has yet to be demonstrated. Other factors, such as discomfort or pain associated with collar use, may also be important, and further work is required in this area.
There were some limitations to this review. The first stage of the selection process was carried out by a single reviewer; however the exclusion criteria were clearly defined and this is unlikely to have influenced the end result. In addition, it was not possible to use a formal tool to assess risk of bias; thus the papers discussed in the results are of varying validity. Nonetheless, in light of the global use of cervical immobilization devices in daily practice, these findings indicate that further scientific scrutiny of the complications described is warranted.
Tissue interface pressure (TIP) is an important factor in the use of cervical immobilisation devices, and to prevent the development of skin ulcers the early removal of a collar is recommended. However, this is not always possible, and where cervical immobilisation is required for a prolonged period of time measures should be taken to reduce the pressure exerted by a collar or similar device. There is no standardised approach to measuring TIP, making it difficult to compare studies and cervical immobilisation devices; however it is reasonable to aim for TIPs of less than 30 mmHg, since this approximates to the capillary closing pressure.
The relationship between cervical collars, the jugular vein and intracranial pressure (ICP) requires further research. Evidence to date demonstrates a strong relationship between cervical collars and altered jugular vein dimensions. This is likely to be the cause of the raised ICP associated with collar application, but has yet to be confirmed experimentally. Research that simultaneously assesses the jugular vein and ICP is therefore required.
New approaches to cervical immobilisation, which avoid these common complications of the collars currently in use, would be welcome.
Search One (tissue interface pressures):
hard cervical collar
#1 OR #2 OR #3 OR #4
#6 OR #7 OR #8 OR #9 0R #10
tissue interface pressure
#12 OR #13 OR #14 OR #15 OR #16 OR #17
5 AND 11 AND 19
Search Two (jugular veins and intracranial pressure):
hard cervical collar
#1 OR #2 OR #3 OR #4
jugular vein diameter
jugular vein measurements
#6 OR #7 OR #8 OR #9 0R #10 OR #11 or #12
5 AND 13
World Health Organisation: Saving Millions of lives: decade of action for road safety 2011-2020. 2011, Geneva: WHO,http://www.who.int/violence_injury_prevention/publications/road_traffic/saving_millions_lives_en.pdf,
American College of Surgeons Committee on Trauma: Advanced Trauma Life Support. 2012, Chicago: American College of Surgeons, 9
el Masri WS: Management of Traumatic Spinal Cord injuries: current standards of care revisited. Adv Clin Neurosci Rehabil. 2010, 10: 37-40.
Walker J: Pressure ulcers in cervical spine immobilisation: a retrospective analysis. J Wound Care. 2012, 21: 323-326.
Jacobson TM, Tescher AN, Miers AG, Downer L: Improving practice: efforts to reduce occipital pressure ulcers. J Nurs Care Qual. 2008, 23: 283-288. 10.1097/01.NCQ.0000324595.29956.90.
Davies G, Deakin C, Wilson A: The effect of a rigid collar on intracranial pressure. Injury. 1996, 27: 647-649. 10.1016/S0020-1383(96)00115-5.
Ho AM, Fung KY, Joynt GM, Karmakar MK, Peng Z: Rigid cervical collar and intracranial pressure of patients with severe head injury. J Trauma-Inj Infect Crit Care. 2002, 53: 1185-1188. 10.1097/00005373-200212000-00028.
Dunham CM, Carter KJ, Castro F, Erickson B: Impact of cervical spine management brain injury on functional survival outcomes in comatose, blunt trauma patients with extremity movement and negative cervical spine CT: application of the Monte Carlo simulation. J Neurotrauma. 2011, 28: 1009-1019. 10.1089/neu.2010.1301.
Mobbs RJ, Stoodley MA, Fuller J: Effect of cervical hard collar on intracranial pressure after head injury. ANZ J Surg. 2002, 72: 389-391. 10.1046/j.1445-2197.2002.02462.x.
Ferguson J, Mardel SN, Beattie TF, Wytch R: Cervical collars: A potential risk to the head-injured patient. Injury. 1993, 24: 454-456. 10.1016/0020-1383(93)90148-Y.
Benger J, Blackham J: Why do we put cervical collars on conscious trauma patients?. Scand J Trauma Resusc Emerg Med. 2009, 17: 44-10.1186/1757-7241-17-44.http://www.sjtrem.com/content/17/1/44,
Voss S, Page M, Benger J: Methods for evaluating cervical range of motion in trauma settings. Scand J Trauma Resusc Emerg Med. 2012, 20: 1-6. 10.1186/1757-7241-20-1.
Liew SCC, Hill DA: Complication of hard cervical collars in multi-trauma patients. Aust N Z J Surg. 1994, 64: 139-140. 10.1111/j.1445-2197.1994.tb02164.x.
Murphy KM, Musiak H, Hanlon M: Patient with occipital pressure ulcer requiring continued cervical collar use. J Wound Ostomy Continence Nurs. 1997, 24: 58-60.
Molano Alvarez E, Murillo Perez MA, Salobral Villegas MT, Dominguez Caballero M, Cuenca Solanas M, Garcia Fuentes C: Pressure sores secondary to immobilization with cervical collar: a complication of acute cervical injury [Spanish] Ulceras por presion secundarias a la inmovilizacion con collarin cervical: una complicacion de la lesion cervical aguda. Enfermeria intensiva / Sociedad Espanola de Enfermeria Intensiva y Unidades Coronarias. 2004, 15: 112-122.
Fisher SV: Proper fitting of the cervical orthosis. Arch Phys Med Rehabil. 1978, 59: 505-507.
Beavis A: Cervical Orthoses. Prosthet Orthot Int. 1989, 13: 6-13.
Plaisier B, Gabram SGA, Schwartz RJ, Jacobs LM: Prospective evaluation of craniofacial pressure in four different cervical orthoses. J Trauma. 1994, 37: 714-720. 10.1097/00005373-199411000-00004.
Blaylock B: Solving the problem of pressure ulcers resulting from cervical collars. Ostomy Wound Manage. 1996, 42: 26-31.
Powers J: A multidisciplinary approach to occipital pressure ulcers related to cervical collars. J Nurs Care Qual. 1997, 12: 46-52.
Chendrasekhar A, Moorman DW, Timberlake GA: An evaluation of the effects of semirigid cervical collars in patients with severe closed head injury. Am Surg. 1998, 64: 604-606.
Black CA, Buderer NM, Blaylock B, Hogan BJ: Comparative study of risk factors for skin breakdown with cervical orthotic devices: PHILADELPHIA and ASPEN. J Trauma Nurs. 1998, 5: 62-66. 10.1097/00043860-199807000-00002.
Tescher AN, Rindflesch AB, Youdas JW, Jacobson TM, Downer LL, Miers AG, Basford JR, Cullinane DC, Stevens SR, Pankratz VS, Decker PA: Range-of-motion restriction and craniofacial tissue-interface pressure from four cervical collars. J Trauma. 2007, 63: 1120-1126. 10.1097/TA.0b013e3180487d0f.
Kolb JC, Summers RL, Galli RL: Cervical collar-induced changes in intracranial pressure. Am J Emerg Med. 1999, 17: 135-137.
Craig GR, Nielsen MS: Rigid cervical collars and intracranial pressure. Intensive Care Med. 1991, 17: 504-505. 10.1007/BF01690778.
Raphael JH, Chotai R: Effects of the cervical collar on cerebrospinal fluid pressure. Anaesthesia. 1994, 49: 437-439.
Porter KM, Allison K: Semirigid cervical collar and intracranial pressure: a change in pre-hospital emphasis might be indicated. Pre-hosp Immediate Care. 1999, 3: 226-228.
Lemyze M, Palud A, Favory R, Mathieu D: Unintentional strangulation by a cervical collar after attempted suicide by hanging. Emerg Med. 2011, 28: 532-10.1136/emj.2010.106625.
Hunt K, Hallworth S, Smith M: The effects of rigid collar placement on intracranial and cerebral perfusion pressures. Anaesthesia. 2001, 56: 511-513. 10.1046/j.1365-2044.2001.02053.x.
Kuhnigk H, Bomke S, Sefrin P: Effect of external cervical spine immobilisation on intracranial pressure [German] Auswirkung der externen halswirbelsaulenimmobilisation auf den intrakraniellen druck. Aktuelle Traumatol. 1993, 23: 350-353.
Stone MB, Tubridy CM, Curran R: The effect of rigid cervical collars on internal jugular vein dimensions. Acad Emerg Med. 2010, 17: 100-102. 10.1111/j.1553-2712.2009.00624.x.
Dunham CM, Brocker BP, Collier BD, Gemmel DJ: Risks associated with magnetic resonance imaging and cervical collar in comatose, blunt trauma patients with negative comprehensive cervical spine computed tomography and no apparent spinal deficit. Crit Care. 2008, 12: R89-10.1186/cc6957.http://ccforum.com/content/12/4/R89,
The authors would like to thank Pauline Shaw (Assistant Librarian) for advice on the search strategy and providing access to papers.
The research team wishes to acknowledge, and thank, all of the ambulance service, fire and rescue service and public volunteers who participated in this research. We are also grateful to Great Western Ambulance Service (now South Western Ambulance Service NHS Foundation Trust) for facilitating and supporting the work.
This research was funded by the National Institute for Health Research Invention for Innovation (i4i) Programme. The views expressed in this report are those of the authors and do not necessarily represent those of the National Institute for Health Research.
The authors declare that they have no competing interests.
AS and SV conducted the review and drafted the manuscript. JB supervised the work, editing the manuscript and providing the clinical context. All authors edited and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
Open Access This article is published under license to BioMed Central Ltd. This is an Open Access article is distributed under the terms of the Creative Commons Attribution License ( https://creativecommons.org/licenses/by/2.0 ), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Sparke, A., Voss, S. & Benger, J. The measurement of tissue interface pressures and changes in jugular venous parameters associated with cervical immobilisation devices: a systematic review. Scand J Trauma Resusc Emerg Med 21, 81 (2013). https://doi.org/10.1186/1757-7241-21-81
- Interface pressure
- Cervical collar
- Pressure ulcers
- Jugular venous pressure
- Intracranial pressure
- Head injury