Does a 4 diagram manual enable laypersons to operate the laryngeal mask supreme®? A pilot study in the manikin
© Schälte et al; licensee BioMed Central Ltd. 2012
Received: 22 November 2011
Accepted: 27 March 2012
Published: 27 March 2012
Bystander resuscitation plays an important role in lifesaving cardiopulmonary resuscitation (CPR). A significant reduction in the "no-flow-time", quantitatively better chest compressions and an improved quality of ventilation can be demonstrated during CPR using supraglottic airway devices (SADs). Previous studies have demonstrated the ability of inexperienced persons to operate SADs after brief instruction. The aim of this pilot study was to determine whether an instruction manual consisting of four diagrams enables laypersons to operate a Laryngeal Mask Supreme® (LMAS) in the manikin.
An instruction manual of four illustrations with speech bubbles displaying the correct use of the LMAS was designed. Laypersons were handed a bag containing a LMAS, a bag mask valve device (BMV), a syringe prefilled with air and the instruction sheet, and were asked to perform and ventilate the manikin as displayed. Time to ventilation was recorded and degree of success evaluated.
A total of 150 laypersons took part. Overall 145 participants (96.7%) inserted the LMAS in the manikin in the right direction. The device was inserted inverted or twisted in 13 (8.7%) attempts. Eight (5.3%) individuals recognized this and corrected the position. Within the first 2 minutes 119 (79.3%) applicants were able to insert the LMAS and provide tidal volumes greater than 150 ml (estimated dead space). Time to insertion and first ventilation was 83.2 ± 29 s. No significant difference related to previous BLS training (P = 0.85), technical education (P = 0.07) or gender could be demonstrated (P = 0.25).
In manikin laypersons could insert LMAS in the correct direction after onsite instruction by a simple manual with a high success rate. This indicates some basic procedural understanding and intellectual transfer in principle. Operating errors (n = 91) were frequently not recognized and corrected (n = 77). Improvements in labeling and the quality of instructional photographs may reduce individual error and may optimize understanding.
Layperson resuscitation plays an important role in providing lifesaving cardiopulmonary resuscitation (CPR) and bridging the interval to the arrival of healthcare professionals. However, only 50% of laypersons actually administer "first aid" in such situations [1, 2]. Reasons given are various and include a lacking sense of personal responsibility when there are many other people "on site", an aversion to strangers' bodily fluids and the percieved risk of infection during "mouth to mouth" ventilation. Individuals may be discouraged from administering CPR by a lack of confidence in their abilities and a fear of doing more harm than good [3–5].
Immediate resuscitation by laypersons and early defibrillation have been identified as key factors positively influencing clinical outcome, survival, and costs [6–8]. In consequence, chest compressions, securing of the airway, initiating (mouth-to-mouth) ventilation, and the use of an automated defibrillator have been implemented in the basic life support guidelines . In the further course of (layperson) CPR securing adequate ventilation is of paramount importance for a full recovery. Mask ventilation requires considerable skill and is associated with the risk of stomach inflation, regurgitation and aspiration. Therefore, it is only recommended for emergency professionals . During CPR a significant reduction in the "no-flow-time", quantitatively better chest compressions and an improved quality of ventilation can be demonstrated using supraglottic airway devices (SADs) [11, 12]. Moreover, unsecured airways during CPR lead to further pulmonary complications. Aspiration is frequently evident in unconscious patients on arrival by ambulance [13–15]. SADs are shown to reduce the incidence of aspiration compared to bag valve mask device ventilation during CPR .
Endotracheal intubation (ETI) remains the standard in pre-hospital airway management, where experienced personnel are present. However, there is controversy as to whether ETI is really the best option for patients when it is performed by paramedics or inexperienced physicians. In these cases, as well as in patients in whom prolonged or multiple intubation attempts are needed, SADs may be the better alternative . SADs have been shown to be easily inserted regardless of level of experience, and properly operated in the manikin as well as in patients in both clinical and preclinical settings [11, 12, 18].
Various training approaches have been investigated. These vary from the traditional (educational session, manikin training and/or practice training, practice) to more compact and minimalistic concepts (i.e. instruction by telephone, educational movie, 3 min demonstration) [19–22].
The aim of this study was to determine whether a short instruction manual consisting of four diagrams only enables laypersons to operate a Laryngeal Mask Supreme® (LMAS) in a manikin, as well as to analyze potential pitfalls and the transfer of given information.
The institutional review board waived the requirement for written informed consent, as no potential harm to the study participants was involved. No personal data except age, academic affiliation and prior "first-aid" knowledge was collected. All subjects agreed to be evaluated anonymously for scientific and educational purposes. Prerequisites for inclusion were the lack of any previous medical education (i.e. physician, nurse, paramedic) other than a BLS course, and an age of 18 or older. Applicants were recruited at the RWTH Aachen University (R heinisch-W estfälische-T echnische-H ochschule Aachen) campus (Audimax & Kármán Auditorium). Experimental data were recorded "on-site" in February and March 2011.
A resuscitation scenario with a manikin (Ambu M MegaCode W®, Ambu GmbH, Bad Nauheim, Germany) was prepared. The experimental scene was separated by partition walls.
Applicants were approached personally and asked to participate in a scientific trial investigating a new alternative to "mouth-to-mouth" ventilation in a dummy and, after agreement, were handed a standardized instruction sheet:
"Behind the wall you will find an unconscious person. This person has stopped breathing. You are responsible for their VENTILATION. In this setting you should use the BOXED DEVICES next to the head. Do not perform "mouth-to-mouth" ventilation. Open the bag and proceed as displayed on the instruction sheet".
Participants then entered the experimental scene and proceeded. Time was recorded starting when the bag was opened and stopped either after ventilation was correctly initiated or the trial was ended by the applicant or - after 4 min - by the investigators. The correct insertion of the LMAS, cuff inflation, connection and compression of the BMV were judged. Multiple compressions of the BMV were allowed as displayed in the manual. In addition, we recorded the number of insertions completed within 2 minutes of the start of the trial. After the trial applicants were interviewed and asked their opinions of the materials, instructions, and their understanding of BMV ventilation. Moreover, they were asked whether they believe that following this instruction sheet and this single trial they would be able to operate the LMAS in a real resuscitation. A tidal volume of > 500 ml for ventilation was judged as sufficient according to the ERC guidelines. Tidal volumes of between 150 ml and 500 ml were judged as "ERC insufficient" but ventilation and tidal volumes < 150 ml were judged as insufficient.
Statistical analysis was performed using SAS (Statistical Analyses System), (SAS Institute GmbH, Heidelberg, Germany). A success rate of 95% was expected [19, 21]. The power of the study was calculated with a significance level, α = 0.05. A power of 80% results in a sample size of 120. In total 150 study subjects were included to compensate for possible dropouts. The power calculation was performed using nQuery Advisor® Version 7.0 (Statistical Solutions, Saugus, MA, USA).
A Chi-square test was used to calculate statistical differences in success rate with respect to gender, previous BLS training, and studying in the field of engineering. T-test was used to calculate statistical differences between time of insertion and age or sex. Correlation was calculated by regression analysis. Data are presented as means ± standard deviation until stated otherwise. A P < 0.05 indicated statistical significance.
23% of applicants complained about problems related to their understanding of the instruction sheet.
A total of 63% of participants reported to be confident in using the LMAS in a real life resuscitation. 33% thought that they would become confident in operating the LMAS after further training, and only 2% stated they would decline the use of LMAS as lay responders.
In this pilot study we evaluated a simple 4-picture instruction manual designed to enable laypersons to operate the LMAS during BLS without any further verbal guidance or "hands-on" training. Overall laypersons were able operate the LMAS with a success rate of 79.3% in the manikin.
This is one of the few trials investigating "genuine" laypersons and their performance operating an SAD "on-site" without any prior training . We focus on specifics of their performance, and expose the pitfalls and potential lack of clarity when transferring pictures into action.
Most of the previous trials published have involved medical, paramedical, nursing students and junior doctors, designating them as "lay persons" [23–28]. Thus, bias resulting from a general interest in the field as well as related medical procedural knowledge cannot be ruled out.
Traditional CPR training is usually in the form of an instructor directed demonstration, frequently followed by brief "hands-on" exercises. Success rates inserting SADs were shown to be as high as > 90% in the manikin and > 80% in patients [23, 26, 29, 30]. As a result, over the past years alternative methods of instruction using newer more media based approaches have become a major topic of research. After scripted telephone instructions, a brief demonstration only, or a video-clip demonstration, success rates of 80% - 95% have been demonstrated [19–21]. In this trial using a four diagram illustrated instruction manual we show that 96.7% of the 150 applicants inserted the LMAS in the right direction. This implicates at least a basically correct transfer of information provided in the diagrams. However, in the 13 participants inserting the device rotated, only 8 recognized and corrected the position. Improved labeling (e.g. similar colors or legible print) and more "close-up" pictures might lead to better understanding. Additional verbal instructions might have solved these problems .
We found the depth of insertion one of the major problems in operating the LMAS in this context. 61 of the LMAS (40.7%) were not inserted at the correct depth as indicated. After recognizing the problem and correcting it, 41 LMAS (27%) remained malpositioned with regard to depth. This might be explained by two factors. First, in the manikin and despite lubrication, friction between two synthetic materials may create the need for more force than is expected by participants. Despite close resemblances even high-fidelity manikins are unable to recreate the feel and finer aspects of human airway anatomy . Second, the landmarks and system of labels chosen are not sufficiently intuitive for laypersons. Nevertheless in 29 of the remaining 41 LMAS inserted not deeply enough ventilation could be provided, though this may possibly be related to characteristics of the Manikin itself. Failing to tilt the manikin's head was the second most obvious mistake made during individual trials (20%). In the 5 cases of impossible ventilation the manikin's head was tilted in anteflexion and not positioned neutral or in retroflexion. In general, insertion of SADs is recommended in a neutral or mildly reclined position but does not require a fully reclined head [29, 30, 32].
Problems with the cuff or its inflation, i.e. an incomplete or omitted inflation, were noted during 21 individual trials (and resulted to some extent in insufficient ventilation (n = 9) of the manikin). None of the participants recognized or corrected this problem. Kurola et al. recently identified cuff related problems as the major hurdles in operating the laryngeal tube following brief written instructions . The authors concluded that special attention should be given to avoiding improper cuff inflation and to the proper depth of insertion, and recommend training by a professional instructor.
Alternatives worth discussing are a self-expanding cuff or an SAD without an inflatable cuff, e.g. the I-Gel laryngeal mask (I-Gel) [25, 33–37]. Eliminating the need to first connect a syringe and then inflate the cuff would eliminate two potential sources of error and thus improve time to ventilation.
Female and male participants performed equally regarding time to successful ventilation in this trial. The prevalence of BLS training in women is hypothesized to be lower in the literature . In this trial no differences between BLS education and gender could be observed.
The time from start of the scenario to definitive ventilation (one measure of "performance") should be interpreted with caution. In comparison to recent trials investigating different training modalities (traditional, verbal, video, telephone or combined), in our trial the time to ventilation was prolonged [19–22]. Participants were instructed "on-site" and needed time to read the visual instructions and identify the devices before taking action. A mean of 83 s seems rather long in this context and differs up to 55 s from the time to ventilation when instructions were given before the start of the trial by which time substantial learning has already occurred . Nevertheless, this is the new approach tested in this study. The starting of resuscitation within the first 2 or 3 minutes following cardiac arrest is compatible with good neurological outcomes, hence this approach may be a step toward better compliance of laypersons with the demand for bystander resuscitation. "On-site" instruction using supraglottic airways may be a useful tool in layperson CPR in multi-bystander scenarios or whenever one of the bystanders is familiar with operating SADs and may correct or improve the initial operator's attempt at insertion .
The RWTH Aachen University was founded as a technical university, with a majority of male students. Schools and faculties of technical and economical engineering still dominate. In this historical context the allocation of participants and the majority of male participants can be explained. We demonstrate that being an engineer, implying a more developed level of technical understanding, did not correlate with a higher success rate compared to non- engineering faculties.
Some limitations need to be discussed. Results presented are obtained in a manikin model and cannot be simply transferred to the patient. The time to insert an SAD in a manikin can be significantly different from that in a patient . Moreover, insertion of an SAD and obtaining a seal is shown to be significantly more difficult in patients than in manikins and is associated with lower success rates and a longer time of insertion . Accordingly we simply note our combined endpoint "time to insertion and successful ventilation" and refrain from further discussion in favor of presenting our procedural findings. One point should be emphasized. This is a feasibility study investigating the efficacy of a standardized instruction manual, and any potential errors transferred into individual practice. Therefore a standardized environment with robust materials - as provided by the manikin - is required . It is well known that no one manikin performs best for SAD insertion. Studying the performance of SADs therefore requires careful selection . Finally, by choosing students we are focused on a highly educated group compared to the general population. It remains speculative as to whether groups of different levels of education would perform differently.
In the manikin retention of skills in operating SADs were shown to be high [18, 24, 39, 40]. Having first aid training increased the likelihood of intervention and of owning a first aid kit or pocket mask .
It remains speculative whether, if supraglottic airway devices were implemented in BLS training, after a certain period of time without training, an illustrated manual may improve laypersons' performance or their willingness to provide (advanced) airway management.
In this pilot study 119 (79.3%) laypersons were able to insert the LMAS and provide tidal volumes greater than 150 ml in the manikin with "on-site" illustrated instructions. Visualizing and understanding the procedure, i.e. inflating the cuff and inserting the LMAS deep enough and in the right direction were identified as the major obstacles. Most faults could have been verbally corrected leading to an even higher success rate. Future improvements in labeling and in the quality of the diagrams may lead to faster and better performance. These findings support the implementation of SADs in BLS training analogously to the implementation of AEDs.
- Handley AJ, Koster R, Monsieurs K, Perkins GD, Davies S, Bossaert L, European Resuscitation Council: European Resuscitation Council guidelines for resuscitation 2005. Section 2. Adult basic life support and use of automated external defibrillators. Resuscitation. 2005, 67 (Suppl 1): S7-S23.View ArticlePubMedGoogle Scholar
- Wiese CH, Bartels U, Bahr J, Graf BM: Kurs Lebensrettende Sofortmaßnahmen, Überprüfung der theoretischen Kenntnisse von Teilnehmern. Notfall Rettungsmed. 2006, 9: 597-603. 10.1007/s10049-006-0852-9.View ArticleGoogle Scholar
- Burghofer K, Schlechtriemen T, Lackner CK: Konsequenzen aus der Altruismusforschung für die Ausbildung in Erster Hilfe. Notfall Rettungsmed. 2005, 8: 408-411. 10.1007/s10049-005-0762-2.View ArticleGoogle Scholar
- Baubin M: Laienreanimation ohne Mund-zu-Mund-Beatmung?. Anaesthesist. 2007, 56: 897-898. 10.1007/s00101-007-1245-2.View ArticleGoogle Scholar
- Kreimeier U, Dirks B, Arntz HR, Bahr J, Goldschmidt P, Roessler M, Sasse M, Toursarkissian M: Stellenwert der Beatmung bei der Laienreanimation, Bestätigung der Gültigkeit der European Resuscitation Council (ERC) Guidelines 2005". Notfall Rettungsmed. 2005, 11: 340-343.View ArticleGoogle Scholar
- Bur A, Kittler H, Sterz F, Holzer M, Eisenburger P, Oschatz E, Kofler J, Laggner AN: Effects of bystander first aid, defibrillation and advanced life support on neurologic outcome and hospital costs in patients after ventricular fibrillation cardiac arrest. Intensive Care Med. 2001, 27: 1474-1480. 10.1007/s001340101045.View ArticlePubMedGoogle Scholar
- Herlitz J, Engdahl J, Svensson L, Angquist KA, Young M, Holmberg S: Factors associated with an increased chance of survival among patients suffering from an out-of-hospital cardiac arrest in a national perspective in Sweden. Am Heart J. 2005, 149: 61-66. 10.1016/j.ahj.2004.07.014.View ArticlePubMedGoogle Scholar
- Bahr J, Panzer W, Klingler H: Herz-Lungen-Wiederbelebung durch Ersthelfer - Einige Ergebnisse und Folgen aus dem Göttinger Pilotprojekt. Anasthesiol Intensivmed Notfallmed Schmerzther. 2001, 36: 573-579. 10.1055/s-2001-17261.View ArticlePubMedGoogle Scholar
- Wenzel V, Russo SG, Arntz HR, Bahr J, Baubin MA, Böttiger BW, Dirks B, Kreimeier U, Fries M, Eich C: Kommentar zu den Leitlinien 2010 zur kardiopulmonalen Reanimation des European Resuscitation Council. Anaesthesist. 2010, 59: 1105-1123. 10.1007/s00101-010-1820-9.View ArticlePubMedGoogle Scholar
- Heuer JF, Barwing J, Eich C, Quintel M, Crozier TA, Roessler M: Initial ventilation through laryngeal tube instead of face mask in out-of-hospital cardiopulmonary arrest is effective and safe. Eur J Emerg Med. 2010, 17: 10-15. 10.1097/MEJ.0b013e32832d852a.View ArticlePubMedGoogle Scholar
- Wiese CH, Semmel T, Müller JU, Bahr J, Ocker H, Graf BM: The use of the laryngeal tube disposable (LT-D) by paramedics during out-of-hospital resuscitation. Resuscitation. 2009, 80: 194-198. 10.1016/j.resuscitation.2008.08.023.View ArticlePubMedGoogle Scholar
- Deakin D, Nolan JP, Soar J, Sunde K, Koster RW, Smith GB, Perkins GD: Erweiterte Reanimationsmaßnahmen für Erwachsene („advanced life support"). Notfall Rettungsmed. 2010, 18: 559-620.View ArticleGoogle Scholar
- Wenzel V, Lindner KH, Prengel AW: Ventilation during cardio-pulmonaryresuscitation (CPR). A literature study and analysis of ventilation strategies. Anaesthesist. 1997, 46: 133-141. 10.1007/s001010050383.View ArticlePubMedGoogle Scholar
- Cuttino M: Management of ventilation during resuscitation. Contemporary cardiology: cardiopulmonary resuscitation. Edited by: Ornato JP, Peberdy MA. Totowa, NJ: Humana Press Inc., S, 95-110.
- Bowman FP, Menegazzi JJ, Check BD, Duckett TM: Lower esophageal sphincter pressure during prolonged cardiac arrest and resuscitation. Ann Emerg Med. 1995, 26: 216-219. 10.1016/S0196-0644(95)70154-0.View ArticlePubMedGoogle Scholar
- Piepho T, Thierbach A, Werner C: Supraglottische Beatmungshilfen in der Notfallmedizin. Notfall Rettungsmed. 2005, 123-128.Google Scholar
- Schalk B, Byhahn C, Fausel F, Egner A, Oberndörfer D, Walcher F, Latasch L: Out-of-hospital airway management by paramedics and emergency physicians using laryngeal tubes. Resuscitation. 2010, 81: 323-326. 10.1016/j.resuscitation.2009.11.007.View ArticlePubMedGoogle Scholar
- Ruetzler K, Roessler B, Potura L, Priemayr A, Robak O, Schuster E, Frass M: Performance and skill retention of intubation by paramedics using seven different airway devices - A manikin study. Resuscitation. 2011, 82: 593-597. 10.1016/j.resuscitation.2011.01.008.View ArticlePubMedGoogle Scholar
- Schälte G, Stoppe C, Aktas M, Coburn M, Rex S, Schwarz M, Rossaint R, Zoremba N: Laypersons can successfully place supraglottic airways with 3 minutes of training. A comparison of four different devices in the manikin. Scand J Trauma Resusc Emerg Med. 2011,Google Scholar
- Beauchamp G, Phrampus P, Guyette FX: Simulated rescue airway use by laypersons with scripted telephonic instruction. Resuscitation. 2009, 80: 925-929. 10.1016/j.resuscitation.2009.04.030.View ArticlePubMedGoogle Scholar
- Jokela J, Nurmi J, Grenzwuerker HV, Castren M: Laryngeal tube and intubating laryngeal mask insertion in a manikin by first-responder trainees after a short video-clip demonstration. Prehosp Disaster Med. 2009, 24: 63-66.PubMedGoogle Scholar
- Kurola J, Paakkonen H, Kettunen T, Laakso JP, Gorski J, Silfvast T: Feasibility of written instructions in airway management training of laryngealtube. Scand J Trauma Resusc Emerg Med. 2011, 10: 19-56.Google Scholar
- Timmermann A, Cremer S, Heuer J, Braun U, Graf BM, Russo SG: Laryngeal mask LMA Supreme. Application by medical personnel inexperienced in airway management. Anaesthesist. 2008, 57: 970-975. 10.1007/s00101-008-1425-8.View ArticlePubMedGoogle Scholar
- Bickenbach J, Schälte G, Beckers S, Fries M, Derwall M, Rossaint R: The intuitive use of laryngeal airway tools by first year medical students. BMC Emerg Med. 2009, 9: 18-23. 10.1186/1471-227X-9-18.PubMed CentralView ArticlePubMedGoogle Scholar
- Fischer H, Hochbrugger E, Fast A, Hager H, Steinlechner B, Koinig H, Eisenburger P, Frantal S, Greif R: Performance of supraglottic airway devices and 12 month skill retention: a randomized controlled study with manikins. Resuscitation. 2011, 82: 326-331. 10.1016/j.resuscitation.2010.11.014.View ArticlePubMedGoogle Scholar
- Howes BW, Wharton NM, Gibbison B, Cook TM: LMA Supreme insertion by novices in manikins and patients. Anaesthesia. 2010, 65: 343-347. 10.1111/j.1365-2044.2010.06262.x.View ArticlePubMedGoogle Scholar
- Coulson A, Brimacombe J, Keller C, Wiseman L, Ingham T, Cheung D, Popwycz L, Hall B: A comparison of the ProSeal and classic laryngeal mask airways for airway management by inexperienced personnel after manikin-only training. Anaesth Intensive Care. 2003, 31: 286-289.PubMedGoogle Scholar
- Yardy N, Hancox D, Strang T: A comparison of two airway aids for emergency use by unskilled personnel. The Combitube and laryngeal mask. Anaesthesia. 1999, 54: 181-183. 10.1046/j.1365-2044.1999.00651.x.View ArticlePubMedGoogle Scholar
- Cook TM, Gatward JJ, Handel J, Hardy R, Thompson C, Srivastava R, Clarke PA: Evaluation of the LMA Supreme in 100 non-paralysed patients. Anaesthesia. 2009, 64: 555-562. 10.1111/j.1365-2044.2008.05824.x.View ArticlePubMedGoogle Scholar
- Verghese C, Ramaswamy B: LMA-Supreme - a new single-use LMA with gastric access: a report on its clinical efficacy. Br J Anaesth. 2008, 101: 405-410. 10.1093/bja/aen174.View ArticlePubMedGoogle Scholar
- Rai MR, Popat MT: Evaluation of airway equipment: man or manikin?. Anaesthesia. 2011, 66 (1): 1-3. 10.1111/j.1365-2044.2010.06567.x.View ArticlePubMedGoogle Scholar
- LMA International, Laryngeal Mask Company Ldt: LMA Supreme Instructions for Use. 2010, [http://www.lmaco.com/pwpcontrol.php?pwpID=5108]Google Scholar
- Noor Zairul M, Khairul Faizi A: Comparison of the VBM laryngeal tube and laryngeal mask airway for ventilation during manual in-line neck stabilisation. Singapore Med J. 2006, 47: 892-896.PubMedGoogle Scholar
- Intersurgical Ltd: i-Gel - Supraglottic Airway Device with Non-inflatable Cuff. 2010, [http://www.i-gel.com/]Google Scholar
- Castle N, Owen R, Hann M, Naidoo R, Reeves D: Assessment of the speed and ease of insertion of three supraglottic airway devices by paramedics: a manikin study. Emerg Med J. 2010, 27: 860-863. 10.1136/emj.2009.084343.View ArticlePubMedGoogle Scholar
- Wharton NM, Gibbison B, Gabbott DA, Haslam GM, Muchatuta N, Cook TM: I-gel insertion by novices in manikins and patients. Anaesthesia. 2008, 63: 991-995. 10.1111/j.1365-2044.2008.05542.x.View ArticlePubMedGoogle Scholar
- Chew EE, Hashim NH, Wang CY: Randomised comparison of the LMA Supreme with the I-Gel in spontaneously breathing anaesthetised adult patients. Anaesth Intensive Care. 2010, 38: 1018-1022.PubMedGoogle Scholar
- Jackson KM, Cook TM: Evaluation of four airway training mainikins as patient simulators for the insertion of eight types of supraglottic airway devices. Anaesthesia. 2007, 62: 388-393. 10.1111/j.1365-2044.2007.04983.x.View ArticlePubMedGoogle Scholar
- Devitt JH, Kurrek MM, Cohen MM, Cleave-Hogg D: The validity of performance assessments using simulation. Anesthesiology. 2001, 95: 36-42. 10.1097/00000542-200107000-00011.View ArticlePubMedGoogle Scholar
- Okazaki Y, Kusunoki S, Kawamoto M: Out-of-hospital tracheal intubation practice by advanced level ambulance crews and reeducation program for skill retention: survey of emergency life saving technicians. Masui. 2011, 60: 757-762.PubMedGoogle Scholar
- Arbon P, Hayes J, Woodman R: First aid and harm minimization for victims of road trauma: a population study. Prehosp Disaster Med. 2011, 26: 276-282. 10.1017/S1049023X11006522.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.